-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170426_115257_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.647227 to 38.838135
-each separately from 0s to 38s and from 472981us to 838135us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.825465 in 1 packets - #/No: 18/18
      22.364576 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-26T11:53:00.647.227.000 to 2017-04-26T11:53:38.838.135.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.825465 in 1 packets - #/No: 18/18
      22.364576 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9335439 to 39.9334055
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999417 in 1 packets - #/No: 18/18
      21.9999199 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      BF -> 34.0°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170426_115257_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.5807699 to 2.5810589
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8254070  -- OK for SL --
   MIU time from 2017-04-26T11:53:00.647.227.000 to 2017-04-26T11:53:00.647.516.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.825465

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4064659 to 18.4067549
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.3644950  -- OK for SL --
   MIU time from 2017-04-26T11:53:16.472.981.000 to 2017-04-26T11:53:16.473.270.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.364576

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.7712499 to 40.7715389
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-26T11:53:38.837.846.000 to 2017-04-26T11:53:38.838.135.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-26T11:53:00.647.086.750
   Max SL corrected packet time: 2017-04-26T11:53:38.837.994.750
   Min SL corrected XO rel time: 0.6472284
   Max SL corrected XO rel time: 38.8381364
   TM values from -22 to 15 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-26T11:53:00.647.086.750 (XO: 0.6472284, t=35.6°C)  Rank: 00004
   TM values: Max =    14  Min =   -22  Ave =    -4.37  Std =     5.53
   Min TM value:-22 at an offset of 89.325 usec -> 2017-04-26T11:53:00.647.176.075 (XO: 0.6473177)
   Max TM value:14 at an offset of 163.226 usec -> 2017-04-26T11:53:00.647.249.975 (XO: 0.6473916)
   Max dTM/dt [1/us]::1.15e+02 at an offset of 251.401 usec -> 2017-04-26T11:53:00.647.338.150 (XO: 0.6474798)
   Min dTM/dt [1/us]::-1.17e+02 at an offset of 215.576 usec -> 2017-04-26T11:53:00.647.302.325 (XO: 0.6474440)

 - Block number: 2 starting at 2017-04-26T11:53:16.472.840.750 (XO: 16.4729820, t=35.6°C)  Rank: 00004
   TM values: Max =    14  Min =   -22  Ave =    -4.39  Std =     5.52
   Min TM value:-22 at an offset of 101.363 usec -> 2017-04-26T11:53:16.472.942.112 (XO: 16.4730834)
   Max TM value:14 at an offset of 141.688 usec -> 2017-04-26T11:53:16.472.982.438 (XO: 16.4731237)
   Max dTM/dt [1/us]::1.14e+02 at an offset of 145.851 usec -> 2017-04-26T11:53:16.472.986.600 (XO: 16.4731279)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 190.676 usec -> 2017-04-26T11:53:16.473.031.425 (XO: 16.4731727)

 - Block number: 3 starting at 2017-04-26T11:53:38.837.705.750 (XO: 38.8378474, t=35.6°C)  Rank: 00004
   TM values: Max =    15  Min =   -21  Ave =    -4.49  Std =     5.08
   Min TM value:-21 at an offset of 123.500 usec -> 2017-04-26T11:53:38.837.829.250 (XO: 38.8379709)
   Max TM value:15 at an offset of 122.613 usec -> 2017-04-26T11:53:38.837.828.362 (XO: 38.8379700)
   Max dTM/dt [1/us]::1.22e+02 at an offset of 158.713 usec -> 2017-04-26T11:53:38.837.864.463 (XO: 38.8380061)
   Min dTM/dt [1/us]::-1.34e+02 at an offset of 123.050 usec -> 2017-04-26T11:53:38.837.828.800 (XO: 38.8379705)
