-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170425_133906_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.055190 to 30.073588
-each separately from 0s to 30s and from 55190us to 195166us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.139398 in 1 packets - #/No: 18/18
      14.878133 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-25T13:39:09.055.190.000 to 2017-04-25T13:39:39.073.588.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.139398 in 1 packets - #/No: 18/18
      14.878133 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4374241 to 31.4373154
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999457 in 1 packets - #/No: 18/18
      14.9999456 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      BF -> 34.0°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170425_133906_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4926154 to 1.4929044
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1393430  -- OK for SL --
   MIU time from 2017-04-25T13:39:09.055.190.000 to 2017-04-25T13:39:09.055.479.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.139398

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6322474 to 16.6325364
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8780790  -- OK for SL --
   MIU time from 2017-04-25T13:39:24.194.877.000 to 2017-04-25T13:39:24.195.166.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.878133

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5106154 to 31.5109044
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-25T13:39:39.073.299.000 to 2017-04-25T13:39:39.073.588.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-25T13:39:09.055.049.750
   Max SL corrected packet time: 2017-04-25T13:39:39.073.447.750
   Min SL corrected XO rel time: 0.0551915
   Max SL corrected XO rel time: 30.0735893
   TM values from -24 to 12 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-25T13:39:09.055.049.750 (XO: 0.0551915, t=35.6°C)  Rank: 00003
   TM values: Max =    12  Min =   -24  Ave =    -4.45  Std =     4.61
   Min TM value:-24 at an offset of 156.101 usec -> 2017-04-25T13:39:09.055.205.850 (XO: 0.0553476)
   Max TM value:12 at an offset of 137.200 usec -> 2017-04-25T13:39:09.055.186.950 (XO: 0.0553287)
   Max dTM/dt [1/us]::1.06e+02 at an offset of 144.238 usec -> 2017-04-25T13:39:09.055.193.988 (XO: 0.0553357)
   Min dTM/dt [1/us]::-1.03e+02 at an offset of 149.488 usec -> 2017-04-25T13:39:09.055.199.238 (XO: 0.0553410)

 - Block number: 2 starting at 2017-04-25T13:39:24.194.736.750 (XO: 15.1948783, t=35.6°C)  Rank: 00003
   TM values: Max =    12  Min =   -21  Ave =    -4.50  Std =     5.13
   Min TM value:-21 at an offset of 170.926 usec -> 2017-04-25T13:39:24.194.907.675 (XO: 15.1950492)
   Max TM value:12 at an offset of 53.100 usec -> 2017-04-25T13:39:24.194.789.850 (XO: 15.1949314)
   Max dTM/dt [1/us]::9.40e+01 at an offset of 183.538 usec -> 2017-04-25T13:39:24.194.920.288 (XO: 15.1950618)
   Min dTM/dt [1/us]::-9.60e+01 at an offset of 117.175 usec -> 2017-04-25T13:39:24.194.853.925 (XO: 15.1949955)

 - Block number: 3 starting at 2017-04-25T13:39:39.073.158.750 (XO: 30.0733003, t=35.6°C)  Rank: 00003
   TM values: Max =    12  Min =   -19  Ave =    -4.47  Std =     4.66
   Min TM value:-19 at an offset of 103.788 usec -> 2017-04-25T13:39:39.073.262.537 (XO: 30.0734041)
   Max TM value:12 at an offset of 110.513 usec -> 2017-04-25T13:39:39.073.269.262 (XO: 30.0734108)
   Max dTM/dt [1/us]::8.90e+01 at an offset of 119.300 usec -> 2017-04-25T13:39:39.073.278.050 (XO: 30.0734196)
   Min dTM/dt [1/us]::-9.40e+01 at an offset of 165.876 usec -> 2017-04-25T13:39:39.073.324.625 (XO: 30.0734661)
