-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170420_113434_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.386784 to 44.598073
-each separately from 4s to 44s and from 90537us to 598073us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.703464 in 1 packets - #/No: 18/18
      29.506958 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-20T11:34:41.386.784.000 to 2017-04-20T11:35:21.598.073.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.703464 in 1 packets - #/No: 18/18
      29.506958 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.4385058 to 45.4383546
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.9999582 in 1 packets - #/No: 18/18
      28.9998906 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170420_113434_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      18 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.8252897 to 5.8255787
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7034230  -- OK for SL --
   MIU time from 2017-04-20T11:34:41.386.784.000 to 2017-04-20T11:34:41.387.073.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.703464

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5290017 to 16.5292907
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.5068470  -- OK for SL --
   MIU time from 2017-04-20T11:34:52.090.537.000 to 2017-04-20T11:34:52.090.826.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.506958

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.0361377 to 46.0364267
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-20T11:35:21.597.784.000 to 2017-04-20T11:35:21.598.073.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-20T11:34:41.386.643.750
   Max SL corrected packet time: 2017-04-20T11:35:21.597.932.750
   Min SL corrected XO rel time: 4.3867854
   Max SL corrected XO rel time: 44.5980752
   TM values from -40 to 31 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-20T11:34:41.386.643.750 (XO: 4.3867854, t=34.0°C)  Rank: 00018
   TM values: Max =    31  Min =   -40  Ave =    -4.41  Std =    15.35
   Min TM value:-40 at an offset of 142.288 usec -> 2017-04-20T11:34:41.386.786.038 (XO: 4.3869277)
   Max TM value:31 at an offset of 117.163 usec -> 2017-04-20T11:34:41.386.760.912 (XO: 4.3869025)
   Max dTM/dt [1/us]::2.92e+02 at an offset of 125.975 usec -> 2017-04-20T11:34:41.386.769.725 (XO: 4.3869113)
   Min dTM/dt [1/us]::-3.01e+02 at an offset of 120.625 usec -> 2017-04-20T11:34:41.386.764.375 (XO: 4.3869060)

 - Block number: 2 starting at 2017-04-20T11:34:52.090.396.750 (XO: 15.0905380, t=34.0°C)  Rank: 00006
   TM values: Max =    22  Min =   -31  Ave =    -4.45  Std =     8.14
   Min TM value:-31 at an offset of 152.176 usec -> 2017-04-20T11:34:52.090.548.925 (XO: 15.0906902)
   Max TM value:22 at an offset of 137.826 usec -> 2017-04-20T11:34:52.090.534.575 (XO: 15.0906759)
   Max dTM/dt [1/us]::1.81e+02 at an offset of 104.138 usec -> 2017-04-20T11:34:52.090.500.887 (XO: 15.0906422)
   Min dTM/dt [1/us]::-1.86e+02 at an offset of 141.288 usec -> 2017-04-20T11:34:52.090.538.038 (XO: 15.0906793)

 - Block number: 3 starting at 2017-04-20T11:35:21.597.643.750 (XO: 44.5977862, t=34.0°C)  Rank: 00007
   TM values: Max =    21  Min =   -31  Ave =    -4.45  Std =     8.94
   Min TM value:-31 at an offset of 136.438 usec -> 2017-04-20T11:35:21.597.780.188 (XO: 44.5979226)
   Max TM value:21 at an offset of 127.925 usec -> 2017-04-20T11:35:21.597.771.675 (XO: 44.5979141)
   Max dTM/dt [1/us]::1.88e+02 at an offset of 170.126 usec -> 2017-04-20T11:35:21.597.813.875 (XO: 44.5979563)
   Min dTM/dt [1/us]::-2.13e+02 at an offset of 134.613 usec -> 2017-04-20T11:35:21.597.778.363 (XO: 44.5979208)
