-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170419_212456_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.236676 to 44.905033
-each separately from 1s to 44s and from 236676us to 973690us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.736436 in 1 packets - #/No: 18/18
      20.931054 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-19T21:25:00.236.676.000 to 2017-04-19T21:25:43.905.033.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.736436 in 1 packets - #/No: 18/18
      20.931054 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4371120 to 45.4369492
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.9999167 in 1 packets - #/No: 18/18
      20.9999205 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170419_212456_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      21 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.6737879 to 2.6740769
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.7363510  -- OK for SL --
   MIU time from 2017-04-19T21:25:00.236.676.000 to 2017-04-19T21:25:00.236.965.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.736436

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.4104279 to 25.4107169
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9309750  -- OK for SL --
   MIU time from 2017-04-19T21:25:22.973.401.000 to 2017-04-19T21:25:22.973.690.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.931054

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      17 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.3416919 to 46.3419809
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-19T21:25:43.904.744.000 to 2017-04-19T21:25:43.905.033.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-19T21:25:00.236.535.750
   Max SL corrected packet time: 2017-04-19T21:25:43.904.892.750
   Min SL corrected XO rel time: 1.2366768
   Max SL corrected XO rel time: 44.9050313
   TM values from -73 to 64 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-19T21:25:00.236.535.750 (XO: 1.2366768, t=35.6°C)  Rank: 00021
   TM values: Max =    58  Min =   -68  Ave =    -4.54  Std =    17.63
   Min TM value:-68 at an offset of 113.038 usec -> 2017-04-19T21:25:00.236.648.787 (XO: 1.2367898)
   Max TM value:58 at an offset of 112.563 usec -> 2017-04-19T21:25:00.236.648.312 (XO: 1.2367894)
   Max dTM/dt [1/us]::9.73e+02 at an offset of 113.350 usec -> 2017-04-19T21:25:00.236.649.100 (XO: 1.2367901)
   Min dTM/dt [1/us]::-9.73e+02 at an offset of 110.963 usec -> 2017-04-19T21:25:00.236.646.712 (XO: 1.2367878)

 - Block number: 2 starting at 2017-04-19T21:25:22.973.260.750 (XO: 23.9734029, t=35.6°C)  Rank: 00012
   TM values: Max =    49  Min =   -56  Ave =    -4.45  Std =    14.38
   Min TM value:-56 at an offset of 260.801 usec -> 2017-04-19T21:25:22.973.521.550 (XO: 23.9736637)
   Max TM value:49 at an offset of 66.738 usec -> 2017-04-19T21:25:22.973.327.487 (XO: 23.9734696)
   Max dTM/dt [1/us]::7.38e+02 at an offset of 119.150 usec -> 2017-04-19T21:25:22.973.379.900 (XO: 23.9735220)
   Min dTM/dt [1/us]::-6.68e+02 at an offset of 119.075 usec -> 2017-04-19T21:25:22.973.379.825 (XO: 23.9735220)

 - Block number: 3 starting at 2017-04-19T21:25:43.904.603.750 (XO: 44.9047423, t=35.6°C)  Rank: 00017
   TM values: Max =    64  Min =   -73  Ave =    -4.52  Std =    17.73
   Min TM value:-73 at an offset of 118.050 usec -> 2017-04-19T21:25:43.904.721.800 (XO: 44.9048604)
   Max TM value:64 at an offset of 120.850 usec -> 2017-04-19T21:25:43.904.724.600 (XO: 44.9048632)
   Max dTM/dt [1/us]::1.12e+03 at an offset of 120.813 usec -> 2017-04-19T21:25:43.904.724.562 (XO: 44.9048632)
   Min dTM/dt [1/us]::-1.02e+03 at an offset of 120.875 usec -> 2017-04-19T21:25:43.904.724.625 (XO: 44.9048632)
