-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170417_055751_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.419757 to 34.560128
-each separately from 4s to 34s and from 382684us to 560128us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.962638 in 1 packets - #/No: 18/18
      5.176866 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-17T05:57:58.419.757.000 to 2017-04-17T05:58:28.560.128.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.962638 in 1 packets - #/No: 18/18
      5.176866 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.4306960 to 35.4305821
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.9999051 in 1 packets - #/No: 18/18
      4.9999810 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170417_055751_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.8504528 to 5.8507418
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.9625430  -- OK for SL --
   MIU time from 2017-04-17T05:57:58.419.757.000 to 2017-04-17T05:57:58.420.046.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.962638

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      18 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.8132848 to 30.8135738
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.1768470  -- OK for SL --
   MIU time from 2017-04-17T05:58:23.382.684.000 to 2017-04-17T05:58:23.382.973.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.176866

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      19 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.9904208 to 35.9907098
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-17T05:58:28.559.839.000 to 2017-04-17T05:58:28.560.128.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-17T05:57:58.419.616.750
   Max SL corrected packet time: 2017-04-17T05:58:28.559.987.750
   Min SL corrected XO rel time: 4.4197584
   Max SL corrected XO rel time: 34.5601299
   TM values from -47 to 38 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-17T05:57:58.419.616.750 (XO: 4.4197584, t=34.0°C)  Rank: 00011
   TM values: Max =    30  Min =   -38  Ave =    -4.45  Std =    12.06
   Min TM value:-38 at an offset of 152.738 usec -> 2017-04-17T05:57:58.419.769.488 (XO: 4.4199111)
   Max TM value:30 at an offset of 118.213 usec -> 2017-04-17T05:57:58.419.734.962 (XO: 4.4198766)
   Max dTM/dt [1/us]::5.64e+02 at an offset of 146.763 usec -> 2017-04-17T05:57:58.419.763.513 (XO: 4.4199052)
   Min dTM/dt [1/us]::-5.64e+02 at an offset of 136.776 usec -> 2017-04-17T05:57:58.419.753.525 (XO: 4.4198952)

 - Block number: 2 starting at 2017-04-17T05:58:23.382.543.750 (XO: 29.3826852, t=34.0°C)  Rank: 00018
   TM values: Max =    36  Min =   -44  Ave =    -4.44  Std =    13.55
   Min TM value:-44 at an offset of 169.676 usec -> 2017-04-17T05:58:23.382.713.425 (XO: 29.3828548)
   Max TM value:36 at an offset of 159.688 usec -> 2017-04-17T05:58:23.382.703.438 (XO: 29.3828448)
   Max dTM/dt [1/us]::6.63e+02 at an offset of 159.651 usec -> 2017-04-17T05:58:23.382.703.400 (XO: 29.3828448)
   Min dTM/dt [1/us]::-6.85e+02 at an offset of 149.651 usec -> 2017-04-17T05:58:23.382.693.400 (XO: 29.3828348)

 - Block number: 3 starting at 2017-04-17T05:58:28.559.698.750 (XO: 34.5598409, t=34.0°C)  Rank: 00019
   TM values: Max =    38  Min =   -47  Ave =    -4.35  Std =    15.31
   Min TM value:-47 at an offset of 170.726 usec -> 2017-04-17T05:58:28.559.869.475 (XO: 34.5600116)
   Max TM value:38 at an offset of 104.363 usec -> 2017-04-17T05:58:28.559.803.112 (XO: 34.5599452)
   Max dTM/dt [1/us]::7.18e+02 at an offset of 117.963 usec -> 2017-04-17T05:58:28.559.816.712 (XO: 34.5599588)
   Min dTM/dt [1/us]::-7.22e+02 at an offset of 165.651 usec -> 2017-04-17T05:58:28.559.864.400 (XO: 34.5600065)
