-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170413_120731_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.009215 to 41.613662
-each separately from 0s to 41s and from 9215us to 613662us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.138787 in 1 packets - #/No: 18/18
      19.464793 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-13T12:07:34.009.215.000 to 2017-04-13T12:08:15.613.662.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.138787 in 1 packets - #/No: 18/18
      19.464793 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4347492 to 42.4345931
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.9999163 in 1 packets - #/No: 18/18
      18.9999276 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170413_120731_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4439651 to 1.4442541
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1387030  -- OK for SL --
   MIU time from 2017-04-13T12:07:34.009.215.000 to 2017-04-13T12:07:34.009.504.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.138787

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.5829571 to 23.5832461
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4647190  -- OK for SL --
   MIU time from 2017-04-13T12:07:56.148.291.000 to 2017-04-13T12:07:56.148.580.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.464793

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.0479651 to 43.0482541
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-13T12:08:15.613.373.000 to 2017-04-13T12:08:15.613.662.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-13T12:07:34.009.074.750
   Max SL corrected packet time: 2017-04-13T12:08:15.613.521.750
   Min SL corrected XO rel time: 0.0092159
   Max SL corrected XO rel time: 41.6136633
   TM values from -24 to 16 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-13T12:07:34.009.074.750 (XO: 0.0092159, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -20  Ave =    -4.48  Std =     4.76
   Min TM value:-20 at an offset of 159.113 usec -> 2017-04-13T12:07:34.009.233.863 (XO: 0.0093750)
   Max TM value:11 at an offset of 156.938 usec -> 2017-04-13T12:07:34.009.231.688 (XO: 0.0093729)
   Max dTM/dt [1/us]::1.00e+02 at an offset of 156.451 usec -> 2017-04-13T12:07:34.009.231.200 (XO: 0.0093724)
   Min dTM/dt [1/us]::-1.07e+02 at an offset of 172.401 usec -> 2017-04-13T12:07:34.009.247.150 (XO: 0.0093883)

 - Block number: 2 starting at 2017-04-13T12:07:56.148.150.750 (XO: 22.1482922, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -21  Ave =    -4.44  Std =     5.67
   Min TM value:-21 at an offset of 116.375 usec -> 2017-04-13T12:07:56.148.267.125 (XO: 22.1484085)
   Max TM value:13 at an offset of 95.900 usec -> 2017-04-13T12:07:56.148.246.650 (XO: 22.1483881)
   Max dTM/dt [1/us]::1.34e+02 at an offset of 116.700 usec -> 2017-04-13T12:07:56.148.267.450 (XO: 22.1484089)
   Min dTM/dt [1/us]::-1.29e+02 at an offset of 48.950 usec -> 2017-04-13T12:07:56.148.199.700 (XO: 22.1483411)

 - Block number: 3 starting at 2017-04-13T12:08:15.613.232.750 (XO: 41.6133743, t=34.0°C)  Rank: 00004
   TM values: Max =    16  Min =   -24  Ave =    -4.32  Std =     6.22
   Min TM value:-24 at an offset of 96.550 usec -> 2017-04-13T12:08:15.613.329.300 (XO: 41.6134709)
   Max TM value:16 at an offset of 144.701 usec -> 2017-04-13T12:08:15.613.377.450 (XO: 41.6135190)
   Max dTM/dt [1/us]::1.45e+02 at an offset of 118.113 usec -> 2017-04-13T12:08:15.613.350.862 (XO: 41.6134924)
   Min dTM/dt [1/us]::-1.41e+02 at an offset of 128.250 usec -> 2017-04-13T12:08:15.613.361.000 (XO: 41.6135026)
