-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170413_030243_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.343792 to 44.382593
-each separately from 6s to 44s and from 87795us to 382593us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.743714 in 1 packets - #/No: 18/18
      29.294220 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-13T03:02:52.343.792.000 to 2017-04-13T03:03:30.382.593.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.743714 in 1 packets - #/No: 18/18
      29.294220 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4380951 to 45.4379513
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999659 in 1 packets - #/No: 18/18
      28.9998903 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170413_030243_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.7818875 to 7.7821765
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7436790  -- OK for SL --
   MIU time from 2017-04-13T03:02:52.343.792.000 to 2017-04-13T03:02:52.344.081.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.743714

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5258555 to 16.5261445
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.2941110  -- OK for SL --
   MIU time from 2017-04-13T03:03:01.087.795.000 to 2017-04-13T03:03:01.088.084.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.294220

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      14 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.8202555 to 45.8205445
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-13T03:03:30.382.304.000 to 2017-04-13T03:03:30.382.593.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-13T03:02:52.343.651.750
   Max SL corrected packet time: 2017-04-13T03:03:30.382.452.750
   Min SL corrected XO rel time: 6.3437937
   Max SL corrected XO rel time: 44.3825910
   TM values from -40 to 30 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-13T03:02:52.343.651.750 (XO: 6.3437937, t=34.0°C)  Rank: 00012
   TM values: Max =    30  Min =   -37  Ave =    -4.40  Std =     9.10
   Min TM value:-37 at an offset of 121.638 usec -> 2017-04-13T03:02:52.343.773.387 (XO: 6.3439153)
   Max TM value:30 at an offset of 124.113 usec -> 2017-04-13T03:02:52.343.775.862 (XO: 6.3439178)
   Max dTM/dt [1/us]::4.88e+02 at an offset of 146.713 usec -> 2017-04-13T03:02:52.343.798.463 (XO: 6.3439404)
   Min dTM/dt [1/us]::-4.91e+02 at an offset of 124.150 usec -> 2017-04-13T03:02:52.343.775.900 (XO: 6.3439179)

 - Block number: 2 starting at 2017-04-13T03:03:01.087.654.750 (XO: 15.0877948, t=34.0°C)  Rank: 00009
   TM values: Max =    25  Min =   -34  Ave =    -4.33  Std =     8.56
   Min TM value:-34 at an offset of 113.150 usec -> 2017-04-13T03:03:01.087.767.900 (XO: 15.0879080)
   Max TM value:25 at an offset of 93.825 usec -> 2017-04-13T03:03:01.087.748.575 (XO: 15.0878887)
   Max dTM/dt [1/us]::4.52e+02 at an offset of 113.363 usec -> 2017-04-13T03:03:01.087.768.112 (XO: 15.0879082)
   Min dTM/dt [1/us]::-4.17e+02 at an offset of 113.788 usec -> 2017-04-13T03:03:01.087.768.537 (XO: 15.0879086)

 - Block number: 3 starting at 2017-04-13T03:03:30.382.163.750 (XO: 44.3823020, t=34.0°C)  Rank: 00014
   TM values: Max =    27  Min =   -40  Ave =    -4.39  Std =    10.70
   Min TM value:-40 at an offset of 187.176 usec -> 2017-04-13T03:03:30.382.350.925 (XO: 44.3824892)
   Max TM value:27 at an offset of 125.300 usec -> 2017-04-13T03:03:30.382.289.050 (XO: 44.3824273)
   Max dTM/dt [1/us]::5.17e+02 at an offset of 124.750 usec -> 2017-04-13T03:03:30.382.288.500 (XO: 44.3824268)
   Min dTM/dt [1/us]::-5.30e+02 at an offset of 143.051 usec -> 2017-04-13T03:03:30.382.306.800 (XO: 44.3824451)
