-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170413_010139_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.097241 to 31.397727
-each separately from 0s to 31s and from 97241us to 764548us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.666729 in 1 packets - #/No: 18/18
      15.632890 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-13T01:01:42.097.241.000 to 2017-04-13T01:02:13.397.727.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.666729 in 1 packets - #/No: 18/18
      15.632890 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4301926 to 32.4300763
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999437 in 1 packets - #/No: 18/18
      15.9999400 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170413_010139_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5274341 to 1.5277231
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6666710  -- OK for SL --
   MIU time from 2017-04-13T01:01:42.097.241.000 to 2017-04-13T01:01:42.097.530.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.666729

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1943941 to 17.1946831
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6328310  -- OK for SL --
   MIU time from 2017-04-13T01:01:57.764.259.000 to 2017-04-13T01:01:57.764.548.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.632890

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.8275141 to 32.8278031
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-13T01:02:13.397.438.000 to 2017-04-13T01:02:13.397.727.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-13T01:01:42.097.100.750
   Max SL corrected packet time: 2017-04-13T01:02:13.397.586.750
   Min SL corrected XO rel time: 0.0972419
   Max SL corrected XO rel time: 31.3977283
   TM values from -24 to 16 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-13T01:01:42.097.100.750 (XO: 0.0972419, t=35.6°C)  Rank: 00003
   TM values: Max =    14  Min =   -23  Ave =    -4.46  Std =     5.14
   Min TM value:-23 at an offset of 125.088 usec -> 2017-04-13T01:01:42.097.225.837 (XO: 0.0973670)
   Max TM value:14 at an offset of 38.000 usec -> 2017-04-13T01:01:42.097.138.750 (XO: 0.0972799)
   Max dTM/dt [1/us]::2.00e+02 at an offset of 11.613 usec -> 2017-04-13T01:01:42.097.112.362 (XO: 0.0972535)
   Min dTM/dt [1/us]::-2.50e+02 at an offset of 16.425 usec -> 2017-04-13T01:01:42.097.117.175 (XO: 0.0972583)

 - Block number: 2 starting at 2017-04-13T01:01:57.764.118.750 (XO: 15.7642606, t=35.6°C)  Rank: 00003
   TM values: Max =    16  Min =   -23  Ave =    -4.39  Std =     5.16
   Min TM value:-23 at an offset of 78.425 usec -> 2017-04-13T01:01:57.764.197.175 (XO: 15.7643391)
   Max TM value:16 at an offset of 195.063 usec -> 2017-04-13T01:01:57.764.313.813 (XO: 15.7644557)
   Max dTM/dt [1/us]::1.89e+02 at an offset of 78.650 usec -> 2017-04-13T01:01:57.764.197.400 (XO: 15.7643393)
   Min dTM/dt [1/us]::-2.25e+02 at an offset of 27.688 usec -> 2017-04-13T01:01:57.764.146.437 (XO: 15.7642883)

 - Block number: 3 starting at 2017-04-13T01:02:13.397.297.750 (XO: 31.3974393, t=35.6°C)  Rank: 00003
   TM values: Max =    16  Min =   -24  Ave =    -4.44  Std =     4.79
   Min TM value:-24 at an offset of 110.188 usec -> 2017-04-13T01:02:13.397.407.937 (XO: 31.3975495)
   Max TM value:16 at an offset of 232.876 usec -> 2017-04-13T01:02:13.397.530.625 (XO: 31.3976722)
   Max dTM/dt [1/us]::2.25e+02 at an offset of 242.338 usec -> 2017-04-13T01:02:13.397.540.088 (XO: 31.3976816)
   Min dTM/dt [1/us]::-2.08e+02 at an offset of 141.251 usec -> 2017-04-13T01:02:13.397.439.000 (XO: 31.3975806)
