-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170412_202915_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.053423 to 33.853666
-each separately from 12s to 33s and from 53423us to 969865us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.915864 in 1 packets - #/No: 18/18
      10.883512 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-12T20:29:30.053.423.000 to 2017-04-12T20:29:51.853.666.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.915864 in 1 packets - #/No: 18/18
      10.883512 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4366250 to 34.4365449
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999619 in 1 packets - #/No: 18/18
      10.9999580 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      C0 -> 35.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170412_202915_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4900488 to 13.4903378
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9158230  -- OK for SL --
   MIU time from 2017-04-12T20:29:30.053.423.000 to 2017-04-12T20:29:30.053.712.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.915864

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.4061608 to 24.4064498
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8834710  -- OK for SL --
   MIU time from 2017-04-12T20:29:40.969.576.000 to 2017-04-12T20:29:40.969.865.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.883512

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.2899208 to 35.2902098
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-12T20:29:51.853.377.000 to 2017-04-12T20:29:51.853.666.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-12T20:29:30.053.282.750
   Max SL corrected packet time: 2017-04-12T20:29:51.853.525.750
   Min SL corrected XO rel time: 12.0534240
   Max SL corrected XO rel time: 33.8536681
   TM values from -74 to 62 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-12T20:29:30.053.282.750 (XO: 12.0534240, t=34.0°C)  Rank: 00013
   TM values: Max =    56  Min =   -66  Ave =    -4.48  Std =    15.63
   Min TM value:-66 at an offset of 109.013 usec -> 2017-04-12T20:29:30.053.391.762 (XO: 12.0535330)
   Max TM value:56 at an offset of 114.288 usec -> 2017-04-12T20:29:30.053.397.037 (XO: 12.0535383)
   Max dTM/dt [1/us]::9.68e+02 at an offset of 216.788 usec -> 2017-04-12T20:29:30.053.499.538 (XO: 12.0536408)
   Min dTM/dt [1/us]::-9.11e+02 at an offset of 216.726 usec -> 2017-04-12T20:29:30.053.499.475 (XO: 12.0536407)

 - Block number: 2 starting at 2017-04-12T20:29:40.969.435.750 (XO: 22.9695777, t=34.0°C)  Rank: 00011
   TM values: Max =    60  Min =   -74  Ave =    -4.51  Std =    16.68
   Min TM value:-74 at an offset of 285.539 usec -> 2017-04-12T20:29:40.969.721.288 (XO: 22.9698632)
   Max TM value:60 at an offset of 140.251 usec -> 2017-04-12T20:29:40.969.576.000 (XO: 22.9697179)
   Max dTM/dt [1/us]::9.04e+02 at an offset of 188.526 usec -> 2017-04-12T20:29:40.969.624.275 (XO: 22.9697662)
   Min dTM/dt [1/us]::-9.54e+02 at an offset of 86.463 usec -> 2017-04-12T20:29:40.969.522.212 (XO: 22.9696641)

 - Block number: 3 starting at 2017-04-12T20:29:51.853.236.750 (XO: 33.8533791, t=34.0°C)  Rank: 00013
   TM values: Max =    62  Min =   -69  Ave =    -4.39  Std =    16.72
   Min TM value:-69 at an offset of 114.275 usec -> 2017-04-12T20:29:51.853.351.025 (XO: 33.8534934)
   Max TM value:62 at an offset of 157.438 usec -> 2017-04-12T20:29:51.853.394.188 (XO: 33.8535366)
   Max dTM/dt [1/us]::9.83e+02 at an offset of 111.975 usec -> 2017-04-12T20:29:51.853.348.725 (XO: 33.8534911)
   Min dTM/dt [1/us]::-9.52e+02 at an offset of 114.238 usec -> 2017-04-12T20:29:51.853.350.987 (XO: 33.8534934)
