-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170412_092323_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.008928 to 30.049123
-each separately from 0s to 30s and from 8928us to 49123us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.011864 in 1 packets - #/No: 18/18
      15.027464 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-12T09:23:26.008.928.000 to 2017-04-12T09:23:56.049.123.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.011864 in 1 packets - #/No: 18/18
      15.027464 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4403499 to 31.4402359
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999430 in 2 packets - #/No: 18/18, 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170412_092323_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4492787 to 1.4495677
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0118070  -- OK for SL --
   MIU time from 2017-04-12T09:23:26.008.928.000 to 2017-04-12T09:23:26.009.217.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.011864

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4613747 to 16.4616637
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0274070  -- OK for SL --
   MIU time from 2017-04-12T09:23:41.021.081.000 to 2017-04-12T09:23:41.021.370.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.027464

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4890707 to 31.4893597
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-12T09:23:56.048.834.000 to 2017-04-12T09:23:56.049.123.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-12T09:23:26.008.787.750
   Max SL corrected packet time: 2017-04-12T09:23:56.048.982.750
   Min SL corrected XO rel time: 0.0089288
   Max SL corrected XO rel time: 30.0491240
   TM values from -21 to 11 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-12T09:23:26.008.787.750 (XO: 0.0089288, t=34.0°C)  Rank: 00003
   TM values: Max =     8  Min =   -20  Ave =    -4.48  Std =     4.74
   Min TM value:-20 at an offset of 134.688 usec -> 2017-04-12T09:23:26.008.922.438 (XO: 0.0090635)
   Max TM value:8 at an offset of 48.600 usec -> 2017-04-12T09:23:26.008.836.350 (XO: 0.0089774)
   Max dTM/dt [1/us]::8.30e+01 at an offset of 294.764 usec -> 2017-04-12T09:23:26.009.082.513 (XO: 0.0092236)
   Min dTM/dt [1/us]::-1.05e+02 at an offset of 186.038 usec -> 2017-04-12T09:23:26.008.973.788 (XO: 0.0091149)

 - Block number: 2 starting at 2017-04-12T09:23:41.020.940.750 (XO: 15.0210819, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -19  Ave =    -4.43  Std =     4.76
   Min TM value:-19 at an offset of 115.638 usec -> 2017-04-12T09:23:41.021.056.387 (XO: 15.0211975)
   Max TM value:11 at an offset of 120.888 usec -> 2017-04-12T09:23:41.021.061.637 (XO: 15.0212028)
   Max dTM/dt [1/us]::9.80e+01 at an offset of 175.326 usec -> 2017-04-12T09:23:41.021.116.075 (XO: 15.0212572)
   Min dTM/dt [1/us]::-1.16e+02 at an offset of 119.850 usec -> 2017-04-12T09:23:41.021.060.600 (XO: 15.0212017)

 - Block number: 3 starting at 2017-04-12T09:23:56.048.693.750 (XO: 30.0488350, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -21  Ave =    -4.45  Std =     4.58
   Min TM value:-21 at an offset of 158.713 usec -> 2017-04-12T09:23:56.048.852.463 (XO: 30.0489937)
   Max TM value:11 at an offset of 169.976 usec -> 2017-04-12T09:23:56.048.863.725 (XO: 30.0490050)
   Max dTM/dt [1/us]::1.11e+02 at an offset of 216.663 usec -> 2017-04-12T09:23:56.048.910.413 (XO: 30.0490516)
   Min dTM/dt [1/us]::-1.00e+02 at an offset of 217.476 usec -> 2017-04-12T09:23:56.048.911.225 (XO: 30.0490525)
