-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170411_091419_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.433276 to 30.029961
-each separately from 1s to 30s and from 29672us to 433565us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.717428 in 1 packets - #/No: 18/18
      1.878390 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-11T09:14:23.433.276.000 to 2017-04-11T09:14:52.029.961.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.717428 in 1 packets - #/No: 18/18
      1.878390 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4769263 to 31.4768173
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.9998986 in 1 packets - #/No: 18/18
      1.9999924 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170411_091419_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.9102021 to 2.9104911
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.7173270  -- OK for SL --
   MIU time from 2017-04-11T09:14:23.433.276.000 to 2017-04-11T09:14:23.433.565.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.717428

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.6278181 to 29.6281071
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.8783830  -- OK for SL --
   MIU time from 2017-04-11T09:14:50.150.993.000 to 2017-04-11T09:14:50.151.282.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.878390

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5064901 to 31.5067791
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-11T09:14:52.029.672.000 to 2017-04-11T09:14:52.029.961.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-11T09:14:23.433.135.750
   Max SL corrected packet time: 2017-04-11T09:14:52.029.820.750
   Min SL corrected XO rel time: 1.4332774
   Max SL corrected XO rel time: 30.0299619
   TM values from -22 to 13 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-11T09:14:23.433.135.750 (XO: 1.4332774, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -22  Ave =    -4.56  Std =     6.37
   Min TM value:-22 at an offset of 79.825 usec -> 2017-04-11T09:14:23.433.215.575 (XO: 1.4333572)
   Max TM value:13 at an offset of 107.938 usec -> 2017-04-11T09:14:23.433.243.687 (XO: 1.4333853)
   Max dTM/dt [1/us]::1.24e+02 at an offset of 141.201 usec -> 2017-04-11T09:14:23.433.276.950 (XO: 1.4334186)
   Min dTM/dt [1/us]::-1.24e+02 at an offset of 131.100 usec -> 2017-04-11T09:14:23.433.266.850 (XO: 1.4334085)

 - Block number: 2 starting at 2017-04-11T09:14:50.150.852.750 (XO: 28.1509938, t=34.0°C)  Rank: 00004
   TM values: Max =     9  Min =   -17  Ave =    -4.49  Std =     4.06
   Min TM value:-17 at an offset of 33.113 usec -> 2017-04-11T09:14:50.150.885.862 (XO: 28.1510269)
   Max TM value:9 at an offset of 12.575 usec -> 2017-04-11T09:14:50.150.865.325 (XO: 28.1510063)
   Max dTM/dt [1/us]::9.80e+01 at an offset of 38.325 usec -> 2017-04-11T09:14:50.150.891.075 (XO: 28.1510321)
   Min dTM/dt [1/us]::-1.17e+02 at an offset of 71.038 usec -> 2017-04-11T09:14:50.150.923.787 (XO: 28.1510648)

 - Block number: 3 starting at 2017-04-11T09:14:52.029.531.750 (XO: 30.0296729, t=34.0°C)  Rank: 00003
   TM values: Max =     9  Min =   -19  Ave =    -4.52  Std =     4.65
   Min TM value:-19 at an offset of 70.975 usec -> 2017-04-11T09:14:52.029.602.725 (XO: 30.0297439)
   Max TM value:9 at an offset of 45.900 usec -> 2017-04-11T09:14:52.029.577.650 (XO: 30.0297188)
   Max dTM/dt [1/us]::1.06e+02 at an offset of 143.088 usec -> 2017-04-11T09:14:52.029.674.838 (XO: 30.0298160)
   Min dTM/dt [1/us]::-1.16e+02 at an offset of 157.338 usec -> 2017-04-11T09:14:52.029.689.088 (XO: 30.0298302)
