-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170409_150039_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.476633 to 37.571895
-each separately from 4s to 37s and from 476633us to 693459us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.216248 in 1 packets - #/No: 18/18
      17.878147 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-09T15:00:46.476.633.000 to 2017-04-09T15:01:19.571.895.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.216248 in 1 packets - #/No: 18/18
      17.878147 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9399604 to 38.9398352
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999430 in 1 packets - #/No: 18/18
      17.9999318 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170409_150039_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      29 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.4165924 to 6.4168814
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.2161910  -- OK for SL --
   MIU time from 2017-04-09T15:00:46.476.633.000 to 2017-04-09T15:00:46.476.922.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.216248

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      26 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6330724 to 21.6333614
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8780790  -- OK for SL --
   MIU time from 2017-04-09T15:01:01.693.170.000 to 2017-04-09T15:01:01.693.459.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.878147

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.5114404 to 39.5117294
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-09T15:01:19.571.606.000 to 2017-04-09T15:01:19.571.895.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-09T15:00:46.476.492.750
   Max SL corrected packet time: 2017-04-09T15:01:19.571.754.750
   Min SL corrected XO rel time: 4.4766338
   Max SL corrected XO rel time: 37.5718964
   TM values from -264 to 220 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-09T15:00:46.476.492.750 (XO: 4.4766338, t=34.0°C)  Rank: 00029
   TM values: Max =   220  Min =  -264  Ave =    -4.50  Std =    16.81
   Min TM value:-264 at an offset of 111.400 usec -> 2017-04-09T15:00:46.476.604.150 (XO: 4.4767452)
   Max TM value:220 at an offset of 111.875 usec -> 2017-04-09T15:00:46.476.604.625 (XO: 4.4767457)
   Max dTM/dt [1/us]::1.60e+03 at an offset of 111.575 usec -> 2017-04-09T15:00:46.476.604.325 (XO: 4.4767454)
   Min dTM/dt [1/us]::-1.94e+03 at an offset of 111.350 usec -> 2017-04-09T15:00:46.476.604.100 (XO: 4.4767452)

 - Block number: 2 starting at 2017-04-09T15:01:01.693.029.750 (XO: 19.6931716, t=34.0°C)  Rank: 00026
   TM values: Max =   111  Min =  -181  Ave =    -4.49  Std =     9.67
   Min TM value:-181 at an offset of 114.938 usec -> 2017-04-09T15:01:01.693.144.687 (XO: 19.6932866)
   Max TM value:111 at an offset of 113.625 usec -> 2017-04-09T15:01:01.693.143.375 (XO: 19.6932853)
   Max dTM/dt [1/us]::2.18e+03 at an offset of 111.638 usec -> 2017-04-09T15:01:01.693.141.387 (XO: 19.6932833)
   Min dTM/dt [1/us]::-1.46e+03 at an offset of 113.275 usec -> 2017-04-09T15:01:01.693.143.025 (XO: 19.6932849)

 - Block number: 3 starting at 2017-04-09T15:01:19.571.465.750 (XO: 37.5716074, t=34.0°C)  Rank: 00006
   TM values: Max =    24  Min =   -34  Ave =    -4.50  Std =     8.75
   Min TM value:-34 at an offset of 151.226 usec -> 2017-04-09T15:01:19.571.616.975 (XO: 37.5717586)
   Max TM value:24 at an offset of 151.263 usec -> 2017-04-09T15:01:19.571.617.013 (XO: 37.5717586)
   Max dTM/dt [1/us]::2.31e+02 at an offset of 0.000 usec -> 2017-04-09T15:01:19.571.465.750 (XO: 37.5716074)
   Min dTM/dt [1/us]::-2.41e+02 at an offset of 151.176 usec -> 2017-04-09T15:01:19.571.616.925 (XO: 37.5717585)
