-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170408_141700_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.086732 to 37.351722
-each separately from 0s to 37s and from 86732us to 413670us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.326360 in 1 packets - #/No: 18/18
      17.937763 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-08T14:17:03.086.732.000 to 2017-04-08T14:17:40.351.722.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.326360 in 1 packets - #/No: 18/18
      17.937763 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4360106 to 38.4358710
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.9999284 in 1 packets - #/No: 18/18
      17.9999320 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      BF -> 34.0°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170408_141700_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5227436 to 1.5230326
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3262870  -- OK for SL --
   MIU time from 2017-04-08T14:17:03.086.732.000 to 2017-04-08T14:17:03.087.021.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.326360

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.8493196 to 20.8496086
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.9376950  -- OK for SL --
   MIU time from 2017-04-08T14:17:22.413.381.000 to 2017-04-08T14:17:22.413.670.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.937763

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.7873036 to 38.7875926
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-08T14:17:40.351.433.000 to 2017-04-08T14:17:40.351.722.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-08T14:17:03.086.591.750
   Max SL corrected packet time: 2017-04-08T14:17:40.351.581.750
   Min SL corrected XO rel time: 0.0867333
   Max SL corrected XO rel time: 37.3517229
   TM values from -24 to 16 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-08T14:17:03.086.591.750 (XO: 0.0867333, t=35.6°C)  Rank: 00003
   TM values: Max =    15  Min =   -24  Ave =    -4.52  Std =     5.83
   Min TM value:-24 at an offset of 152.238 usec -> 2017-04-08T14:17:03.086.743.988 (XO: 0.0868856)
   Max TM value:15 at an offset of 146.788 usec -> 2017-04-08T14:17:03.086.738.538 (XO: 0.0868801)
   Max dTM/dt [1/us]::1.28e+02 at an offset of 147.988 usec -> 2017-04-08T14:17:03.086.739.738 (XO: 0.0868813)
   Min dTM/dt [1/us]::-1.48e+02 at an offset of 151.913 usec -> 2017-04-08T14:17:03.086.743.663 (XO: 0.0868852)

 - Block number: 2 starting at 2017-04-08T14:17:22.413.240.750 (XO: 19.4133822, t=35.6°C)  Rank: 00004
   TM values: Max =    16  Min =   -24  Ave =    -4.40  Std =     6.03
   Min TM value:-24 at an offset of 99.950 usec -> 2017-04-08T14:17:22.413.340.700 (XO: 19.4134821)
   Max TM value:16 at an offset of 115.850 usec -> 2017-04-08T14:17:22.413.356.600 (XO: 19.4134980)
   Max dTM/dt [1/us]::1.33e+02 at an offset of 118.750 usec -> 2017-04-08T14:17:22.413.359.500 (XO: 19.4135009)
   Min dTM/dt [1/us]::-1.41e+02 at an offset of 116.288 usec -> 2017-04-08T14:17:22.413.357.037 (XO: 19.4134985)

 - Block number: 3 starting at 2017-04-08T14:17:40.351.292.750 (XO: 37.3514339, t=35.6°C)  Rank: 00005
   TM values: Max =    11  Min =   -20  Ave =    -4.37  Std =     4.40
   Min TM value:-20 at an offset of 99.900 usec -> 2017-04-08T14:17:40.351.392.650 (XO: 37.3515338)
   Max TM value:11 at an offset of 90.050 usec -> 2017-04-08T14:17:40.351.382.800 (XO: 37.3515240)
   Max dTM/dt [1/us]::1.36e+02 at an offset of 101.800 usec -> 2017-04-08T14:17:40.351.394.550 (XO: 37.3515357)
   Min dTM/dt [1/us]::-1.08e+02 at an offset of 55.325 usec -> 2017-04-08T14:17:40.351.348.075 (XO: 37.3514892)
