-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170408_091405_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.010833 to 38.056990
-each separately from 1s to 38s and from 10833us to 429992us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.418581 in 1 packets - #/No: 18/18
      22.626709 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-08T09:14:09.010.833.000 to 2017-04-08T09:14:46.056.990.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.418581 in 1 packets - #/No: 18/18
      22.626709 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4358021 to 39.4356629
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999473 in 1 packets - #/No: 18/18
      22.9999135 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170408_091405_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4466364 to 2.4469254
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4185270  -- OK for SL --
   MIU time from 2017-04-08T09:14:09.010.833.000 to 2017-04-08T09:14:09.011.122.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.418581

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8654524 to 16.8657414
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6266230  -- OK for SL --
   MIU time from 2017-04-08T09:14:23.429.703.000 to 2017-04-08T09:14:23.429.992.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.626709

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.4923644 to 39.4926534
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-08T09:14:46.056.701.000 to 2017-04-08T09:14:46.056.990.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-08T09:14:09.010.692.750
   Max SL corrected packet time: 2017-04-08T09:14:46.056.849.750
   Min SL corrected XO rel time: 1.0108343
   Max SL corrected XO rel time: 38.0569907
   TM values from -22 to 13 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-08T09:14:09.010.692.750 (XO: 1.0108343, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -19  Ave =    -4.48  Std =     4.79
   Min TM value:-19 at an offset of 173.088 usec -> 2017-04-08T09:14:09.010.865.838 (XO: 1.0110074)
   Max TM value:12 at an offset of 125.138 usec -> 2017-04-08T09:14:09.010.817.887 (XO: 1.0109595)
   Max dTM/dt [1/us]::1.06e+02 at an offset of 164.138 usec -> 2017-04-08T09:14:09.010.856.888 (XO: 1.0109985)
   Min dTM/dt [1/us]::-1.04e+02 at an offset of 95.375 usec -> 2017-04-08T09:14:09.010.788.125 (XO: 1.0109297)

 - Block number: 2 starting at 2017-04-08T09:14:23.429.562.750 (XO: 15.4297046, t=34.0°C)  Rank: 00003
   TM values: Max =     9  Min =   -18  Ave =    -4.52  Std =     4.79
   Min TM value:-18 at an offset of 140.063 usec -> 2017-04-08T09:14:23.429.702.813 (XO: 15.4298447)
   Max TM value:9 at an offset of 133.138 usec -> 2017-04-08T09:14:23.429.695.888 (XO: 15.4298378)
   Max dTM/dt [1/us]::9.50e+01 at an offset of 36.888 usec -> 2017-04-08T09:14:23.429.599.637 (XO: 15.4297415)
   Min dTM/dt [1/us]::-9.80e+01 at an offset of 203.663 usec -> 2017-04-08T09:14:23.429.766.413 (XO: 15.4299083)

 - Block number: 3 starting at 2017-04-08T09:14:46.056.560.750 (XO: 38.0567017, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -22  Ave =    -4.55  Std =     5.39
   Min TM value:-22 at an offset of 129.000 usec -> 2017-04-08T09:14:46.056.689.750 (XO: 38.0568307)
   Max TM value:13 at an offset of 170.926 usec -> 2017-04-08T09:14:46.056.731.675 (XO: 38.0568726)
   Max dTM/dt [1/us]::1.37e+02 at an offset of 199.538 usec -> 2017-04-08T09:14:46.056.760.288 (XO: 38.0569013)
   Min dTM/dt [1/us]::-1.31e+02 at an offset of 119.575 usec -> 2017-04-08T09:14:46.056.680.325 (XO: 38.0568213)
