-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170405_113931_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.059869 to 32.504945
-each separately from 2s to 32s and from 59869us to 579336us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.518889 in 1 packets - #/No: 18/18
      10.925320 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-05T11:39:36.059.869.000 to 2017-04-05T11:40:06.504.945.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.518889 in 1 packets - #/No: 18/18
      10.925320 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4344543 to 33.4343408
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.9999282 in 1 packets - #/No: 18/18
      10.9999583 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170405_113931_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.4943246 to 3.4946136
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5188150  -- OK for SL --
   MIU time from 2017-04-05T11:39:36.059.869.000 to 2017-04-05T11:39:36.060.158.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.518889

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.0134286 to 23.0137176
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9252790  -- OK for SL --
   MIU time from 2017-04-05T11:39:55.579.047.000 to 2017-04-05T11:39:55.579.336.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.925320

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.9389966 to 33.9392856
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-05T11:40:06.504.656.000 to 2017-04-05T11:40:06.504.945.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-05T11:39:36.059.728.750
   Max SL corrected packet time: 2017-04-05T11:40:06.504.804.750
   Min SL corrected XO rel time: 2.0598705
   Max SL corrected XO rel time: 32.5049467
   TM values from -23 to 14 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-05T11:39:36.059.728.750 (XO: 2.0598705, t=34.0°C)  Rank: 00004
   TM values: Max =    14  Min =   -23  Ave =    -4.48  Std =     5.33
   Min TM value:-23 at an offset of 158.501 usec -> 2017-04-05T11:39:36.059.887.250 (XO: 2.0600290)
   Max TM value:14 at an offset of 180.613 usec -> 2017-04-05T11:39:36.059.909.363 (XO: 2.0600511)
   Max dTM/dt [1/us]::1.27e+02 at an offset of 216.713 usec -> 2017-04-05T11:39:36.059.945.463 (XO: 2.0600872)
   Min dTM/dt [1/us]::-1.22e+02 at an offset of 290.614 usec -> 2017-04-05T11:39:36.060.019.363 (XO: 2.0601611)

 - Block number: 2 starting at 2017-04-05T11:39:55.578.906.750 (XO: 21.5790483, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -23  Ave =    -4.43  Std =     5.49
   Min TM value:-23 at an offset of 81.138 usec -> 2017-04-05T11:39:55.578.987.887 (XO: 21.5791294)
   Max TM value:13 at an offset of 113.963 usec -> 2017-04-05T11:39:55.579.020.712 (XO: 21.5791623)
   Max dTM/dt [1/us]::1.11e+02 at an offset of 84.713 usec -> 2017-04-05T11:39:55.578.991.462 (XO: 21.5791330)
   Min dTM/dt [1/us]::-1.18e+02 at an offset of 213.026 usec -> 2017-04-05T11:39:55.579.119.775 (XO: 21.5792613)

 - Block number: 3 starting at 2017-04-05T11:40:06.504.515.750 (XO: 32.5046577, t=34.0°C)  Rank: 00006
   TM values: Max =    11  Min =   -20  Ave =    -4.53  Std =     4.82
   Min TM value:-20 at an offset of 73.075 usec -> 2017-04-05T11:40:06.504.588.825 (XO: 32.5047307)
   Max TM value:11 at an offset of 61.550 usec -> 2017-04-05T11:40:06.504.577.300 (XO: 32.5047192)
   Max dTM/dt [1/us]::1.01e+02 at an offset of 64.463 usec -> 2017-04-05T11:40:06.504.580.212 (XO: 32.5047221)
   Min dTM/dt [1/us]::-1.02e+02 at an offset of 109.100 usec -> 2017-04-05T11:40:06.504.624.850 (XO: 32.5047668)
