-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170401_093925_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.548030 to 30.165993
-each separately from 14s to 30s and from 21072us to 548319us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.472753 in 1 packets - #/No: 18/18
      15.144343 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-01T09:39:42.548.030.000 to 2017-04-01T09:39:58.165.993.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.472753 in 1 packets - #/No: 18/18
      15.144343 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4401094 to 31.4400496
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.9999962 in 1 packets - #/No: 18/18
      14.9999440 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170401_093925_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.9881386 to 15.9884276
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.4727510  -- OK for SL --
   MIU time from 2017-04-01T09:39:42.548.030.000 to 2017-04-01T09:39:42.548.319.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.472753

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4611786 to 16.4614676
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1442870  -- OK for SL --
   MIU time from 2017-04-01T09:39:43.021.072.000 to 2017-04-01T09:39:43.021.361.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.144343

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.6057546 to 31.6060436
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-01T09:39:58.165.704.000 to 2017-04-01T09:39:58.165.993.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-01T09:39:42.547.889.750
   Max SL corrected packet time: 2017-04-01T09:39:58.165.852.750
   Min SL corrected XO rel time: 14.5480313
   Max SL corrected XO rel time: 30.1659946
   TM values from -23 to 18 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-01T09:39:42.547.889.750 (XO: 14.5480313, t=34.0°C)  Rank: 00004
   TM values: Max =    18  Min =   -23  Ave =    -4.42  Std =     4.44
   Min TM value:-23 at an offset of 191.938 usec -> 2017-04-01T09:39:42.548.081.688 (XO: 14.5482232)
   Max TM value:18 at an offset of 259.451 usec -> 2017-04-01T09:39:42.548.149.200 (XO: 14.5482907)
   Max dTM/dt [1/us]::1.43e+02 at an offset of 280.551 usec -> 2017-04-01T09:39:42.548.170.301 (XO: 14.5483118)
   Min dTM/dt [1/us]::-1.44e+02 at an offset of 289.226 usec -> 2017-04-01T09:39:42.548.178.976 (XO: 14.5483205)

 - Block number: 2 starting at 2017-04-01T09:39:43.020.931.750 (XO: 15.0210731, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -22  Ave =    -4.37  Std =     5.38
   Min TM value:-22 at an offset of 179.163 usec -> 2017-04-01T09:39:43.021.110.913 (XO: 15.0212522)
   Max TM value:10 at an offset of 77.913 usec -> 2017-04-01T09:39:43.021.009.662 (XO: 15.0211510)
   Max dTM/dt [1/us]::1.03e+02 at an offset of 77.788 usec -> 2017-04-01T09:39:43.021.009.537 (XO: 15.0211509)
   Min dTM/dt [1/us]::-1.21e+02 at an offset of 134.638 usec -> 2017-04-01T09:39:43.021.066.387 (XO: 15.0212077)

 - Block number: 3 starting at 2017-04-01T09:39:58.165.563.750 (XO: 30.1657056, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -19  Ave =    -4.35  Std =     5.36
   Min TM value:-19 at an offset of 162.963 usec -> 2017-04-01T09:39:58.165.726.713 (XO: 30.1658686)
   Max TM value:12 at an offset of 107.463 usec -> 2017-04-01T09:39:58.165.671.212 (XO: 30.1658131)
   Max dTM/dt [1/us]::1.21e+02 at an offset of 105.550 usec -> 2017-04-01T09:39:58.165.669.300 (XO: 30.1658112)
   Min dTM/dt [1/us]::-1.11e+02 at an offset of 139.563 usec -> 2017-04-01T09:39:58.165.703.313 (XO: 30.1658452)
