-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170325_085822_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.464427 to 30.429170
-each separately from 5s to 30s and from 172093us to 464716us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.707377 in 1 packets - #/No: 18/18
      1.256499 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-25T08:58:30.464.427.000 to 2017-03-25T08:58:55.429.170.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.707377 in 1 packets - #/No: 18/18
      1.256499 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9319747 to 31.9318887
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999175 in 1 packets - #/No: 18/18
      0.9999965 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170325_085822_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.3964016 to 7.3966906
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.7072950  -- OK for SL --
   MIU time from 2017-03-25T08:58:30.464.427.000 to 2017-03-25T08:58:30.464.716.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.707377

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      19 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.1039856 to 31.1042746
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.2564950  -- OK for SL --
   MIU time from 2017-03-25T08:58:54.172.093.000 to 2017-03-25T08:58:54.172.382.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.256499

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      19 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.3607696 to 32.3610586
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-25T08:58:55.428.881.000 to 2017-03-25T08:58:55.429.170.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-25T08:58:30.464.286.750
   Max SL corrected packet time: 2017-03-25T08:58:55.429.029.750
   Min SL corrected XO rel time: 5.4644285
   Max SL corrected XO rel time: 30.4291714
   TM values from -48 to 40 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-25T08:58:30.464.286.750 (XO: 5.4644285, t=35.6°C)  Rank: 00005
   TM values: Max =    17  Min =   -25  Ave =    -4.51  Std =     6.40
   Min TM value:-25 at an offset of 149.076 usec -> 2017-03-25T08:58:30.464.435.825 (XO: 5.4645776)
   Max TM value:17 at an offset of 120.700 usec -> 2017-03-25T08:58:30.464.407.450 (XO: 5.4645492)
   Max dTM/dt [1/us]::2.02e+02 at an offset of 123.513 usec -> 2017-03-25T08:58:30.464.410.262 (XO: 5.4645520)
   Min dTM/dt [1/us]::-2.05e+02 at an offset of 179.176 usec -> 2017-03-25T08:58:30.464.465.925 (XO: 5.4646077)

 - Block number: 2 starting at 2017-03-25T08:58:54.171.952.750 (XO: 29.1720940, t=35.6°C)  Rank: 00019
   TM values: Max =    37  Min =   -46  Ave =    -4.59  Std =    16.77
   Min TM value:-46 at an offset of 122.775 usec -> 2017-03-25T08:58:54.172.075.525 (XO: 29.1722168)
   Max TM value:37 at an offset of 141.963 usec -> 2017-03-25T08:58:54.172.094.712 (XO: 29.1722360)
   Max dTM/dt [1/us]::3.52e+02 at an offset of 131.163 usec -> 2017-03-25T08:58:54.172.083.912 (XO: 29.1722252)
   Min dTM/dt [1/us]::-3.61e+02 at an offset of 195.038 usec -> 2017-03-25T08:58:54.172.147.788 (XO: 29.1722890)

 - Block number: 3 starting at 2017-03-25T08:58:55.428.740.750 (XO: 30.4288824, t=35.6°C)  Rank: 00019
   TM values: Max =    40  Min =   -48  Ave =    -4.60  Std =    15.13
   Min TM value:-48 at an offset of 162.251 usec -> 2017-03-25T08:58:55.428.903.000 (XO: 30.4290446)
   Max TM value:40 at an offset of 178.376 usec -> 2017-03-25T08:58:55.428.919.125 (XO: 30.4290607)
   Max dTM/dt [1/us]::3.68e+02 at an offset of 176.238 usec -> 2017-03-25T08:58:55.428.916.988 (XO: 30.4290586)
   Min dTM/dt [1/us]::-3.72e+02 at an offset of 160.963 usec -> 2017-03-25T08:58:55.428.901.713 (XO: 30.4290433)
