-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170324_084534_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.751867 to 42.458881
-each separately from 8s to 42s and from 458592us to 752156us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.724535 in 1 packets - #/No: 18/18
      26.981612 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-24T08:45:45.751.867.000 to 2017-03-24T08:46:19.458.881.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.724535 in 1 packets - #/No: 18/18
      26.981612 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4254253 to 43.4253074
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999756 in 1 packets - #/No: 18/18
      26.9999065 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170324_084534_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.1772914 to 10.1775804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.7245110  -- OK for SL --
   MIU time from 2017-03-24T08:45:45.751.867.000 to 2017-03-24T08:45:45.752.156.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.724535

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9020914 to 16.9023804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.9815190  -- OK for SL --
   MIU time from 2017-03-24T08:45:52.476.691.000 to 2017-03-24T08:45:52.476.980.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.981612

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.8838994 to 43.8841884
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-24T08:46:19.458.592.000 to 2017-03-24T08:46:19.458.881.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-24T08:45:45.751.726.750
   Max SL corrected packet time: 2017-03-24T08:46:19.458.740.750
   Min SL corrected XO rel time: 8.7518687
   Max SL corrected XO rel time: 42.4588826
   TM values from -30 to 25 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-24T08:45:45.751.726.750 (XO: 8.7518687, t=35.6°C)  Rank: 00005
   TM values: Max =    25  Min =   -30  Ave =    -4.49  Std =     8.99
   Min TM value:-30 at an offset of 205.288 usec -> 2017-03-24T08:45:45.751.932.038 (XO: 8.7520740)
   Max TM value:25 at an offset of 52.550 usec -> 2017-03-24T08:45:45.751.779.300 (XO: 8.7519213)
   Max dTM/dt [1/us]::2.22e+02 at an offset of 119.150 usec -> 2017-03-24T08:45:45.751.845.900 (XO: 8.7519879)
   Min dTM/dt [1/us]::-2.09e+02 at an offset of 186.476 usec -> 2017-03-24T08:45:45.751.913.225 (XO: 8.7520552)

 - Block number: 2 starting at 2017-03-24T08:45:52.476.550.750 (XO: 15.4766921, t=35.6°C)  Rank: 00005
   TM values: Max =    19  Min =   -28  Ave =    -4.66  Std =     7.71
   Min TM value:-28 at an offset of 124.588 usec -> 2017-03-24T08:45:52.476.675.337 (XO: 15.4768167)
   Max TM value:19 at an offset of 61.363 usec -> 2017-03-24T08:45:52.476.612.112 (XO: 15.4767535)
   Max dTM/dt [1/us]::2.04e+02 at an offset of 61.313 usec -> 2017-03-24T08:45:52.476.612.062 (XO: 15.4767534)
   Min dTM/dt [1/us]::-1.92e+02 at an offset of 131.763 usec -> 2017-03-24T08:45:52.476.682.512 (XO: 15.4768239)

 - Block number: 3 starting at 2017-03-24T08:46:19.458.451.750 (XO: 42.4585936, t=35.6°C)  Rank: 00005
   TM values: Max =    18  Min =   -28  Ave =    -4.49  Std =     7.96
   Min TM value:-28 at an offset of 74.113 usec -> 2017-03-24T08:46:19.458.525.862 (XO: 42.4586677)
   Max TM value:18 at an offset of 14.013 usec -> 2017-03-24T08:46:19.458.465.762 (XO: 42.4586076)
   Max dTM/dt [1/us]::1.79e+02 at an offset of 177.938 usec -> 2017-03-24T08:46:19.458.629.688 (XO: 42.4587715)
   Min dTM/dt [1/us]::-1.78e+02 at an offset of 111.975 usec -> 2017-03-24T08:46:19.458.563.725 (XO: 42.4587056)
