-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170323_063004_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.331655 to 31.423260
-each separately from 2s to 31s and from 331655us to 454492us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.122259 in 1 packets - #/No: 18/18
      13.968479 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-23T06:30:09.331.655.000 to 2017-03-23T06:30:38.423.260.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.122259 in 1 packets - #/No: 18/18
      13.968479 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9246428 to 32.9245429
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999483 in 1 packets - #/No: 18/18
      13.9999518 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170323_063004_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2562977 to 4.2565867
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1222070  -- OK for SL --
   MIU time from 2017-03-23T06:30:09.331.655.000 to 2017-03-23T06:30:09.331.944.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.122259

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.3787937 to 19.3790827
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.9684310  -- OK for SL --
   MIU time from 2017-03-23T06:30:24.454.203.000 to 2017-03-23T06:30:24.454.492.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.968479

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.3475137 to 33.3478027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-23T06:30:38.422.971.000 to 2017-03-23T06:30:38.423.260.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-23T06:30:09.331.514.750
   Max SL corrected packet time: 2017-03-23T06:30:38.423.119.750
   Min SL corrected XO rel time: 2.3316560
   Max SL corrected XO rel time: 31.4232613
   TM values from -32 to 23 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-23T06:30:09.331.514.750 (XO: 2.3316560, t=34.0°C)  Rank: 00004
   TM values: Max =    23  Min =   -29  Ave =    -4.47  Std =     6.22
   Min TM value:-29 at an offset of 67.588 usec -> 2017-03-23T06:30:09.331.582.337 (XO: 2.3317236)
   Max TM value:23 at an offset of 93.413 usec -> 2017-03-23T06:30:09.331.608.162 (XO: 2.3317494)
   Max dTM/dt [1/us]::3.66e+02 at an offset of 102.563 usec -> 2017-03-23T06:30:09.331.617.312 (XO: 2.3317586)
   Min dTM/dt [1/us]::-4.11e+02 at an offset of 93.450 usec -> 2017-03-23T06:30:09.331.608.200 (XO: 2.3317495)

 - Block number: 2 starting at 2017-03-23T06:30:24.454.062.750 (XO: 17.4542041, t=34.0°C)  Rank: 00004
   TM values: Max =    20  Min =   -30  Ave =    -4.59  Std =     7.19
   Min TM value:-30 at an offset of 247.988 usec -> 2017-03-23T06:30:24.454.310.738 (XO: 17.4544521)
   Max TM value:20 at an offset of 119.538 usec -> 2017-03-23T06:30:24.454.182.287 (XO: 17.4543237)
   Max dTM/dt [1/us]::3.30e+02 at an offset of 178.301 usec -> 2017-03-23T06:30:24.454.241.050 (XO: 17.4543824)
   Min dTM/dt [1/us]::-3.27e+02 at an offset of 119.575 usec -> 2017-03-23T06:30:24.454.182.325 (XO: 17.4543237)

 - Block number: 3 starting at 2017-03-23T06:30:38.422.830.750 (XO: 31.4229723, t=34.0°C)  Rank: 00004
   TM values: Max =    20  Min =   -32  Ave =    -4.59  Std =     6.70
   Min TM value:-32 at an offset of 138.925 usec -> 2017-03-23T06:30:38.422.969.675 (XO: 31.4231112)
   Max TM value:20 at an offset of 124.550 usec -> 2017-03-23T06:30:38.422.955.300 (XO: 31.4230968)
   Max dTM/dt [1/us]::3.42e+02 at an offset of 85.138 usec -> 2017-03-23T06:30:38.422.915.887 (XO: 31.4230574)
   Min dTM/dt [1/us]::-3.48e+02 at an offset of 96.125 usec -> 2017-03-23T06:30:38.422.926.875 (XO: 31.4230684)
