-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170323_052932_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.112411 to 30.856902
-each separately from 0s to 30s and from 24117us to 856902us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.911417 in 1 packets - #/No: 18/18
      13.832207 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-23T05:29:35.112.411.000 to 2017-03-23T05:30:05.856.902.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.911417 in 1 packets - #/No: 18/18
      13.832207 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4291040 to 31.4290007
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999415 in 1 packets - #/No: 18/18
      12.9999552 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170323_052932_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5415159 to 1.5418049
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9113590  -- OK for SL --
   MIU time from 2017-03-23T05:29:35.112.411.000 to 2017-03-23T05:29:35.112.700.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.911417

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4531639 to 18.4534529
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8321590  -- OK for SL --
   MIU time from 2017-03-23T05:29:52.024.117.000 to 2017-03-23T05:29:52.024.406.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.832207

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.2856119 to 32.2859009
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-23T05:30:05.856.613.000 to 2017-03-23T05:30:05.856.902.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-23T05:29:35.112.270.750
   Max SL corrected packet time: 2017-03-23T05:30:05.856.761.750
   Min SL corrected XO rel time: 0.1124123
   Max SL corrected XO rel time: 30.8569031
   TM values from -26 to 16 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-23T05:29:35.112.270.750 (XO: 0.1124123, t=34.0°C)  Rank: 00004
   TM values: Max =    14  Min =   -26  Ave =    -4.44  Std =     5.74
   Min TM value:-26 at an offset of 149.738 usec -> 2017-03-23T05:29:35.112.420.488 (XO: 0.1125620)
   Max TM value:14 at an offset of 113.813 usec -> 2017-03-23T05:29:35.112.384.562 (XO: 0.1125261)
   Max dTM/dt [1/us]::1.80e+02 at an offset of 145.413 usec -> 2017-03-23T05:29:35.112.416.163 (XO: 0.1125577)
   Min dTM/dt [1/us]::-1.82e+02 at an offset of 164.026 usec -> 2017-03-23T05:29:35.112.434.775 (XO: 0.1125763)

 - Block number: 2 starting at 2017-03-23T05:29:52.023.976.750 (XO: 17.0241185, t=34.0°C)  Rank: 00004
   TM values: Max =    15  Min =   -26  Ave =    -4.50  Std =     5.71
   Min TM value:-26 at an offset of 79.613 usec -> 2017-03-23T05:29:52.024.056.362 (XO: 17.0241981)
   Max TM value:15 at an offset of 123.163 usec -> 2017-03-23T05:29:52.024.099.912 (XO: 17.0242416)
   Max dTM/dt [1/us]::1.86e+02 at an offset of 157.776 usec -> 2017-03-23T05:29:52.024.134.525 (XO: 17.0242763)
   Min dTM/dt [1/us]::-1.84e+02 at an offset of 106.588 usec -> 2017-03-23T05:29:52.024.083.337 (XO: 17.0242251)

 - Block number: 3 starting at 2017-03-23T05:30:05.856.472.750 (XO: 30.8566141, t=34.0°C)  Rank: 00004
   TM values: Max =    16  Min =   -26  Ave =    -4.45  Std =     6.06
   Min TM value:-26 at an offset of 37.625 usec -> 2017-03-23T05:30:05.856.510.375 (XO: 30.8566517)
   Max TM value:16 at an offset of 155.601 usec -> 2017-03-23T05:30:05.856.628.350 (XO: 30.8567697)
   Max dTM/dt [1/us]::1.76e+02 at an offset of 183.826 usec -> 2017-03-23T05:30:05.856.656.575 (XO: 30.8567979)
   Min dTM/dt [1/us]::-1.70e+02 at an offset of 97.988 usec -> 2017-03-23T05:30:05.856.570.737 (XO: 30.8567121)
