-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170322_142132_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.459999 to 44.641107
-each separately from 6s to 44s and from 459999us to 897984us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.437407 in 1 packets - #/No: 18/18
      28.742834 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-22T14:21:41.459.999.000 to 2017-03-22T14:22:19.641.107.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.437407 in 1 packets - #/No: 18/18
      28.742834 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4295158 to 45.4293856
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999691 in 1 packets - #/No: 18/18
      28.9999007 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170322_142132_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8895146 to 7.8898036
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4373750  -- OK for SL --
   MIU time from 2017-03-22T14:21:41.459.999.000 to 2017-03-22T14:21:41.460.288.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.437407

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3271786 to 17.3274676
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.7427350  -- OK for SL --
   MIU time from 2017-03-22T14:21:50.897.695.000 to 2017-03-22T14:21:50.897.984.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.742834

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.0702026 to 46.0704916
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-22T14:22:19.640.818.000 to 2017-03-22T14:22:19.641.107.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-22T14:21:41.459.858.750
   Max SL corrected packet time: 2017-03-22T14:22:19.640.966.750
   Min SL corrected XO rel time: 6.4600004
   Max SL corrected XO rel time: 44.6411071
   TM values from -25 to 15 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-22T14:21:41.459.858.750 (XO: 6.4600004, t=34.0°C)  Rank: 00004
   TM values: Max =    15  Min =   -24  Ave =    -4.55  Std =     5.48
   Min TM value:-24 at an offset of 116.475 usec -> 2017-03-22T14:21:41.459.975.225 (XO: 6.4601169)
   Max TM value:15 at an offset of 120.575 usec -> 2017-03-22T14:21:41.459.979.325 (XO: 6.4601210)
   Max dTM/dt [1/us]::1.01e+02 at an offset of 40.563 usec -> 2017-03-22T14:21:41.459.899.312 (XO: 6.4600410)
   Min dTM/dt [1/us]::-1.25e+02 at an offset of 216.526 usec -> 2017-03-22T14:21:41.460.075.275 (XO: 6.4602169)

 - Block number: 2 starting at 2017-03-22T14:21:50.897.554.750 (XO: 15.8976968, t=34.0°C)  Rank: 00004
   TM values: Max =    12  Min =   -23  Ave =    -4.55  Std =     4.89
   Min TM value:-23 at an offset of 118.925 usec -> 2017-03-22T14:21:50.897.673.675 (XO: 15.8978157)
   Max TM value:12 at an offset of 96.525 usec -> 2017-03-22T14:21:50.897.651.275 (XO: 15.8977933)
   Max dTM/dt [1/us]::1.17e+02 at an offset of 53.700 usec -> 2017-03-22T14:21:50.897.608.450 (XO: 15.8977505)
   Min dTM/dt [1/us]::-1.00e+02 at an offset of 133.500 usec -> 2017-03-22T14:21:50.897.688.250 (XO: 15.8978303)

 - Block number: 3 starting at 2017-03-22T14:22:19.640.677.750 (XO: 44.6408181, t=34.0°C)  Rank: 00004
   TM values: Max =    15  Min =   -25  Ave =    -4.50  Std =     6.21
   Min TM value:-25 at an offset of 130.338 usec -> 2017-03-22T14:22:19.640.808.087 (XO: 44.6409484)
   Max TM value:15 at an offset of 132.913 usec -> 2017-03-22T14:22:19.640.810.662 (XO: 44.6409510)
   Max dTM/dt [1/us]::1.35e+02 at an offset of 257.063 usec -> 2017-03-22T14:22:19.640.934.813 (XO: 44.6410751)
   Min dTM/dt [1/us]::-1.16e+02 at an offset of 104.088 usec -> 2017-03-22T14:22:19.640.781.837 (XO: 44.6409222)
