-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170312_161647_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.691224 to 35.769730
-each separately from 6s to 35s and from 54823us to 769730us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.363310 in 1 packets - #/No: 18/18
      20.714329 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-12T16:16:56.691.224.000 to 2017-03-12T16:17:25.769.730.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.363310 in 1 packets - #/No: 18/18
      20.714329 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9282112 to 36.9281065
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999675 in 1 packets - #/No: 18/18
      19.9999278 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170312_161647_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6194349 to 8.6197239
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3632790  -- OK for SL --
   MIU time from 2017-03-12T16:16:56.691.224.000 to 2017-03-12T16:16:56.691.513.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.363310

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9830029 to 16.9832919
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.7142550  -- OK for SL --
   MIU time from 2017-03-12T16:17:05.054.823.000 to 2017-03-12T16:17:05.055.112.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.714329

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.6975469 to 37.6978359
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-12T16:17:25.769.441.000 to 2017-03-12T16:17:25.769.730.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-12T16:16:56.691.083.750
   Max SL corrected packet time: 2017-03-12T16:17:25.769.589.750
   Min SL corrected XO rel time: 6.6912262
   Max SL corrected XO rel time: 35.7697322
   TM values from -36 to 25 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-12T16:16:56.691.083.750 (XO: 6.6912262, t=35.6°C)  Rank: 00005
   TM values: Max =    25  Min =   -34  Ave =    -4.43  Std =     7.16
   Min TM value:-34 at an offset of 130.600 usec -> 2017-03-12T16:16:56.691.214.350 (XO: 6.6913568)
   Max TM value:25 at an offset of 161.813 usec -> 2017-03-12T16:16:56.691.245.563 (XO: 6.6913880)
   Max dTM/dt [1/us]::3.34e+02 at an offset of 81.863 usec -> 2017-03-12T16:16:56.691.165.612 (XO: 6.6913081)
   Min dTM/dt [1/us]::-3.41e+02 at an offset of 145.088 usec -> 2017-03-12T16:16:56.691.228.838 (XO: 6.6913713)

 - Block number: 2 starting at 2017-03-12T16:17:05.054.682.750 (XO: 15.0548244, t=35.6°C)  Rank: 00004
   TM values: Max =    23  Min =   -32  Ave =    -4.31  Std =     7.71
   Min TM value:-32 at an offset of 194.438 usec -> 2017-03-12T16:17:05.054.877.188 (XO: 15.0550188)
   Max TM value:23 at an offset of 135.888 usec -> 2017-03-12T16:17:05.054.818.637 (XO: 15.0549603)
   Max dTM/dt [1/us]::3.23e+02 at an offset of 135.850 usec -> 2017-03-12T16:17:05.054.818.600 (XO: 15.0549602)
   Min dTM/dt [1/us]::-3.33e+02 at an offset of 87.363 usec -> 2017-03-12T16:17:05.054.770.112 (XO: 15.0549118)

 - Block number: 3 starting at 2017-03-12T16:17:25.769.300.750 (XO: 35.7694432, t=35.6°C)  Rank: 00005
   TM values: Max =    24  Min =   -36  Ave =    -4.46  Std =     7.27
   Min TM value:-36 at an offset of 73.363 usec -> 2017-03-12T16:17:25.769.374.112 (XO: 35.7695165)
   Max TM value:24 at an offset of 112.325 usec -> 2017-03-12T16:17:25.769.413.075 (XO: 35.7695555)
   Max dTM/dt [1/us]::2.87e+02 at an offset of 102.763 usec -> 2017-03-12T16:17:25.769.403.512 (XO: 35.7695459)
   Min dTM/dt [1/us]::-2.90e+02 at an offset of 86.588 usec -> 2017-03-12T16:17:25.769.387.337 (XO: 35.7695298)
