-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170310_073411_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.955295 to 36.020131
-each separately from 11s to 36s and from 19842us to 955584us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.639622 in 1 packets - #/No: 18/18
      17.424347 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-10T07:34:25.955.295.000 to 2017-03-10T07:34:50.020.131.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.639622 in 1 packets - #/No: 18/18
      17.424347 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9316140 to 37.9315279
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999759 in 1 packets - #/No: 18/18
      17.9999380 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170310_073411_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.8869075 to 13.8871965
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.6395990  -- OK for SL --
   MIU time from 2017-03-10T07:34:25.955.295.000 to 2017-03-10T07:34:25.955.584.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.639622

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.5267955 to 20.5270845
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4242870  -- OK for SL --
   MIU time from 2017-03-10T07:34:32.595.206.000 to 2017-03-10T07:34:32.595.495.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.424347

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.9513715 to 37.9516605
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-10T07:34:50.019.842.000 to 2017-03-10T07:34:50.020.131.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-10T07:34:25.955.154.750
   Max SL corrected packet time: 2017-03-10T07:34:50.019.990.750
   Min SL corrected XO rel time: 11.9552968
   Max SL corrected XO rel time: 36.0201327
   TM values from -31 to 22 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-10T07:34:25.955.154.750 (XO: 11.9552968, t=35.6°C)  Rank: 00005
   TM values: Max =    20  Min =   -27  Ave =    -4.56  Std =     6.10
   Min TM value:-27 at an offset of 99.838 usec -> 2017-03-10T07:34:25.955.254.587 (XO: 11.9553967)
   Max TM value:20 at an offset of 124.750 usec -> 2017-03-10T07:34:25.955.279.500 (XO: 11.9554216)
   Max dTM/dt [1/us]::2.31e+02 at an offset of 124.713 usec -> 2017-03-10T07:34:25.955.279.462 (XO: 11.9554216)
   Min dTM/dt [1/us]::-2.20e+02 at an offset of 97.813 usec -> 2017-03-10T07:34:25.955.252.562 (XO: 11.9553947)

 - Block number: 2 starting at 2017-03-10T07:34:32.595.065.750 (XO: 18.5952076, t=35.6°C)  Rank: 00005
   TM values: Max =    18  Min =   -27  Ave =    -4.50  Std =     6.23
   Min TM value:-27 at an offset of 181.413 usec -> 2017-03-10T07:34:32.595.247.163 (XO: 18.5953890)
   Max TM value:18 at an offset of 240.063 usec -> 2017-03-10T07:34:32.595.305.813 (XO: 18.5954477)
   Max dTM/dt [1/us]::2.18e+02 at an offset of 181.451 usec -> 2017-03-10T07:34:32.595.247.200 (XO: 18.5953891)
   Min dTM/dt [1/us]::-2.11e+02 at an offset of 72.888 usec -> 2017-03-10T07:34:32.595.138.637 (XO: 18.5952805)

 - Block number: 3 starting at 2017-03-10T07:34:50.019.701.750 (XO: 36.0198437, t=35.6°C)  Rank: 00005
   TM values: Max =    22  Min =   -31  Ave =    -4.39  Std =     9.10
   Min TM value:-31 at an offset of 108.325 usec -> 2017-03-10T07:34:50.019.810.075 (XO: 36.0199520)
   Max TM value:22 at an offset of 188.288 usec -> 2017-03-10T07:34:50.019.890.038 (XO: 36.0200320)
   Max dTM/dt [1/us]::2.44e+02 at an offset of 130.200 usec -> 2017-03-10T07:34:50.019.831.950 (XO: 36.0199739)
   Min dTM/dt [1/us]::-2.33e+02 at an offset of 99.025 usec -> 2017-03-10T07:34:50.019.800.775 (XO: 36.0199427)
