-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170309_205835_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.650831 to 32.123924
-each separately from 8s to 32s and from 123635us to 651120us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.514040 in 1 packets - #/No: 18/18
      11.958186 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-09T20:58:46.650.831.000 to 2017-03-09T20:59:10.123.924.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.514040 in 1 packets - #/No: 18/18
      11.958186 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4262982 to 33.4262130
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999574 in 2 packets - #/No: 18/18, 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170309_205835_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.0771285 to 10.0774175
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.5139990  -- OK for SL --
   MIU time from 2017-03-09T20:58:46.650.831.000 to 2017-03-09T20:58:46.651.120.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.514040

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.5914165 to 21.5917055
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9581430  -- OK for SL --
   MIU time from 2017-03-09T20:58:58.165.160.000 to 2017-03-09T20:58:58.165.449.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.958186

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.5498485 to 33.5501375
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-09T20:59:10.123.635.000 to 2017-03-09T20:59:10.123.924.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-09T20:58:46.650.690.750
   Max SL corrected packet time: 2017-03-09T20:59:10.123.783.750
   Min SL corrected XO rel time: 8.6508326
   Max SL corrected XO rel time: 32.1239249
   TM values from -58 to 53 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-09T20:58:46.650.690.750 (XO: 8.6508326, t=35.6°C)  Rank: 00012
   TM values: Max =    50  Min =   -58  Ave =    -4.52  Std =    13.61
   Min TM value:-58 at an offset of 81.438 usec -> 2017-03-09T20:58:46.650.772.187 (XO: 8.6509140)
   Max TM value:50 at an offset of 80.025 usec -> 2017-03-09T20:58:46.650.770.775 (XO: 8.6509126)
   Max dTM/dt [1/us]::7.74e+02 at an offset of 122.438 usec -> 2017-03-09T20:58:46.650.813.187 (XO: 8.6509550)
   Min dTM/dt [1/us]::-8.31e+02 at an offset of 106.913 usec -> 2017-03-09T20:58:46.650.797.662 (XO: 8.6509395)

 - Block number: 2 starting at 2017-03-09T20:58:58.165.019.750 (XO: 20.1651615, t=35.6°C)  Rank: 00013
   TM values: Max =    53  Min =   -55  Ave =    -4.51  Std =    12.84
   Min TM value:-55 at an offset of 217.313 usec -> 2017-03-09T20:58:58.165.237.063 (XO: 20.1653788)
   Max TM value:53 at an offset of 114.713 usec -> 2017-03-09T20:58:58.165.134.462 (XO: 20.1652762)
   Max dTM/dt [1/us]::8.63e+02 at an offset of 116.188 usec -> 2017-03-09T20:58:58.165.135.937 (XO: 20.1652777)
   Min dTM/dt [1/us]::-9.89e+02 at an offset of 116.263 usec -> 2017-03-09T20:58:58.165.136.012 (XO: 20.1652778)

 - Block number: 3 starting at 2017-03-09T20:59:10.123.494.750 (XO: 32.1236359, t=35.6°C)  Rank: 00011
   TM values: Max =    40  Min =   -56  Ave =    -4.48  Std =    13.47
   Min TM value:-56 at an offset of 125.525 usec -> 2017-03-09T20:59:10.123.620.275 (XO: 32.1237615)
   Max TM value:40 at an offset of 201.126 usec -> 2017-03-09T20:59:10.123.695.875 (XO: 32.1238371)
   Max dTM/dt [1/us]::8.35e+02 at an offset of 125.563 usec -> 2017-03-09T20:59:10.123.620.312 (XO: 32.1237615)
   Min dTM/dt [1/us]::-8.04e+02 at an offset of 125.500 usec -> 2017-03-09T20:59:10.123.620.250 (XO: 32.1237614)
