-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170308_113854_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.913806 to 38.350738
-each separately from 5s to 38s and from 350449us to 914095us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.537485 in 1 packets - #/No: 18/18
      14.898580 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-08T11:39:02.913.806.000 to 2017-03-08T11:39:35.350.738.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.537485 in 1 packets - #/No: 18/18
      14.898580 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4289092 to 39.4287927
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999365 in 1 packets - #/No: 18/18
      14.9999470 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170308_113854_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.3427137 to 7.3430027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5374230  -- OK for SL --
   MIU time from 2017-03-08T11:39:02.913.806.000 to 2017-03-08T11:39:02.914.095.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.537485

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.8804257 to 24.8807147
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8985270  -- OK for SL --
   MIU time from 2017-03-08T11:39:20.451.580.000 to 2017-03-08T11:39:20.451.869.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.898580

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.7792417 to 39.7795307
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-08T11:39:35.350.449.000 to 2017-03-08T11:39:35.350.738.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-08T11:39:02.913.665.750
   Max SL corrected packet time: 2017-03-08T11:39:35.350.597.750
   Min SL corrected XO rel time: 5.9138077
   Max SL corrected XO rel time: 38.3507393
   TM values from -29 to 20 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-08T11:39:02.913.665.750 (XO: 5.9138077, t=35.6°C)  Rank: 00006
   TM values: Max =    20  Min =   -29  Ave =    -4.52  Std =     7.75
   Min TM value:-29 at an offset of 125.050 usec -> 2017-03-08T11:39:02.913.790.800 (XO: 5.9139327)
   Max TM value:20 at an offset of 149.838 usec -> 2017-03-08T11:39:02.913.815.588 (XO: 5.9139575)
   Max dTM/dt [1/us]::2.22e+02 at an offset of 286.339 usec -> 2017-03-08T11:39:02.913.952.088 (XO: 5.9140940)
   Min dTM/dt [1/us]::-2.15e+02 at an offset of 125.975 usec -> 2017-03-08T11:39:02.913.791.725 (XO: 5.9139337)

 - Block number: 2 starting at 2017-03-08T11:39:20.451.439.750 (XO: 23.4515816, t=35.6°C)  Rank: 00005
   TM values: Max =    18  Min =   -27  Ave =    -4.57  Std =     6.31
   Min TM value:-27 at an offset of 143.926 usec -> 2017-03-08T11:39:20.451.583.675 (XO: 23.4517256)
   Max TM value:18 at an offset of 115.788 usec -> 2017-03-08T11:39:20.451.555.537 (XO: 23.4516974)
   Max dTM/dt [1/us]::1.65e+02 at an offset of 165.651 usec -> 2017-03-08T11:39:20.451.605.400 (XO: 23.4517473)
   Min dTM/dt [1/us]::-1.77e+02 at an offset of 178.776 usec -> 2017-03-08T11:39:20.451.618.525 (XO: 23.4517604)

 - Block number: 3 starting at 2017-03-08T11:39:35.350.308.750 (XO: 38.3504503, t=35.6°C)  Rank: 00006
   TM values: Max =    14  Min =   -23  Ave =    -4.59  Std =     4.95
   Min TM value:-23 at an offset of 50.025 usec -> 2017-03-08T11:39:35.350.358.775 (XO: 38.3505003)
   Max TM value:14 at an offset of 231.801 usec -> 2017-03-08T11:39:35.350.540.550 (XO: 38.3506821)
   Max dTM/dt [1/us]::1.36e+02 at an offset of 193.463 usec -> 2017-03-08T11:39:35.350.502.213 (XO: 38.3506437)
   Min dTM/dt [1/us]::-1.45e+02 at an offset of 161.038 usec -> 2017-03-08T11:39:35.350.469.788 (XO: 38.3506113)
