-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170303_222728_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.566212 to 31.844459
-each separately from 7s to 31s and from 566212us to 844459us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.042025 in 1 packets - #/No: 18/18
      3.235355 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-03T22:27:38.566.212.000 to 2017-03-03T22:28:02.844.459.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.042025 in 1 packets - #/No: 18/18
      3.235355 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9294451 to 32.9293607
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999261 in 1 packets - #/No: 18/18
      2.9999895 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170303_222728_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.4956573 to 9.4959463
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0419510  -- OK for SL --
   MIU time from 2017-03-03T22:27:38.566.212.000 to 2017-03-03T22:27:38.566.501.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.042025

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.5378973 to 30.5381863
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.2353430  -- OK for SL --
   MIU time from 2017-03-03T22:27:59.608.526.000 to 2017-03-03T22:27:59.608.815.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.235355

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.7735293 to 33.7738183
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-03T22:28:02.844.170.000 to 2017-03-03T22:28:02.844.459.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-03T22:27:38.566.071.750
   Max SL corrected packet time: 2017-03-03T22:28:02.844.318.750
   Min SL corrected XO rel time: 7.5662142
   Max SL corrected XO rel time: 31.8444606
   TM values from -38 to 28 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-03T22:27:38.566.071.750 (XO: 7.5662142, t=35.6°C)  Rank: 00008
   TM values: Max =    21  Min =   -31  Ave =    -4.57  Std =     7.22
   Min TM value:-31 at an offset of 119.913 usec -> 2017-03-03T22:27:38.566.191.662 (XO: 7.5663341)
   Max TM value:21 at an offset of 131.288 usec -> 2017-03-03T22:27:38.566.203.037 (XO: 7.5663455)
   Max dTM/dt [1/us]::2.53e+02 at an offset of 215.951 usec -> 2017-03-03T22:27:38.566.287.700 (XO: 7.5664301)
   Min dTM/dt [1/us]::-2.77e+02 at an offset of 122.500 usec -> 2017-03-03T22:27:38.566.194.250 (XO: 7.5663367)

 - Block number: 2 starting at 2017-03-03T22:27:59.608.385.750 (XO: 28.6085282, t=35.6°C)  Rank: 00008
   TM values: Max =    28  Min =   -38  Ave =    -4.43  Std =     8.48
   Min TM value:-38 at an offset of 125.413 usec -> 2017-03-03T22:27:59.608.511.162 (XO: 28.6086536)
   Max TM value:28 at an offset of 127.675 usec -> 2017-03-03T22:27:59.608.513.425 (XO: 28.6086559)
   Max dTM/dt [1/us]::2.87e+02 at an offset of 239.888 usec -> 2017-03-03T22:27:59.608.625.638 (XO: 28.6087681)
   Min dTM/dt [1/us]::-2.81e+02 at an offset of 292.464 usec -> 2017-03-03T22:27:59.608.678.213 (XO: 28.6088207)

 - Block number: 3 starting at 2017-03-03T22:28:02.844.029.750 (XO: 31.8441716, t=35.6°C)  Rank: 00008
   TM values: Max =    27  Min =   -33  Ave =    -4.47  Std =     8.27
   Min TM value:-33 at an offset of 182.251 usec -> 2017-03-03T22:28:02.844.212.000 (XO: 31.8443538)
   Max TM value:27 at an offset of 115.913 usec -> 2017-03-03T22:28:02.844.145.662 (XO: 31.8442875)
   Max dTM/dt [1/us]::2.71e+02 at an offset of 222.138 usec -> 2017-03-03T22:28:02.844.251.888 (XO: 31.8443937)
   Min dTM/dt [1/us]::-2.96e+02 at an offset of 221.251 usec -> 2017-03-03T22:28:02.844.251.000 (XO: 31.8443928)
