-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170302_211253_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.124371 to 30.384776
-each separately from 11s to 30s and from 124371us to 384776us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.193677 in 1 packets - #/No: 18/18
      15.065861 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-02T21:13:07.124.371.000 to 2017-03-02T21:13:26.384.776.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.193677 in 1 packets - #/No: 18/18
      15.065861 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.4259595 to 31.4258921
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.9999858 in 1 packets - #/No: 18/18
      14.9999468 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170302_211253_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.5503308 to 12.5506198
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.1936630  -- OK for SL --
   MIU time from 2017-03-02T21:13:07.124.371.000 to 2017-03-02T21:13:07.124.660.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.193677

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7442828 to 16.7445718
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0658070  -- OK for SL --
   MIU time from 2017-03-02T21:13:11.318.337.000 to 2017-03-02T21:13:11.318.626.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.065861

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.8103788 to 31.8106678
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-02T21:13:26.384.487.000 to 2017-03-02T21:13:26.384.776.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-02T21:13:07.124.230.750
   Max SL corrected packet time: 2017-03-02T21:13:26.384.635.750
   Min SL corrected XO rel time: 11.1243717
   Max SL corrected XO rel time: 30.3847770
   TM values from -37 to 34 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-02T21:13:07.124.230.750 (XO: 11.1243717, t=35.6°C)  Rank: 00007
   TM values: Max =    25  Min =   -34  Ave =    -4.50  Std =     8.13
   Min TM value:-34 at an offset of 116.213 usec -> 2017-03-02T21:13:07.124.346.962 (XO: 11.1244880)
   Max TM value:25 at an offset of 93.600 usec -> 2017-03-02T21:13:07.124.324.350 (XO: 11.1244653)
   Max dTM/dt [1/us]::3.05e+02 at an offset of 38.413 usec -> 2017-03-02T21:13:07.124.269.162 (XO: 11.1244102)
   Min dTM/dt [1/us]::-3.19e+02 at an offset of 154.601 usec -> 2017-03-02T21:13:07.124.385.350 (XO: 11.1245263)

 - Block number: 2 starting at 2017-03-02T21:13:11.318.196.750 (XO: 15.3183386, t=35.6°C)  Rank: 00006
   TM values: Max =    23  Min =   -32  Ave =    -4.48  Std =     7.80
   Min TM value:-32 at an offset of 129.250 usec -> 2017-03-02T21:13:11.318.326.000 (XO: 15.3184679)
   Max TM value:23 at an offset of 128.638 usec -> 2017-03-02T21:13:11.318.325.387 (XO: 15.3184673)
   Max dTM/dt [1/us]::3.02e+02 at an offset of 88.838 usec -> 2017-03-02T21:13:11.318.285.587 (XO: 15.3184275)
   Min dTM/dt [1/us]::-3.03e+02 at an offset of 87.950 usec -> 2017-03-02T21:13:11.318.284.700 (XO: 15.3184266)

 - Block number: 3 starting at 2017-03-02T21:13:26.384.346.750 (XO: 30.3844880, t=35.6°C)  Rank: 00006
   TM values: Max =    34  Min =   -37  Ave =    -4.50  Std =     9.42
   Min TM value:-37 at an offset of 142.675 usec -> 2017-03-02T21:13:26.384.489.425 (XO: 30.3846307)
   Max TM value:34 at an offset of 40.075 usec -> 2017-03-02T21:13:26.384.386.825 (XO: 30.3845281)
   Max dTM/dt [1/us]::4.12e+02 at an offset of 123.088 usec -> 2017-03-02T21:13:26.384.469.837 (XO: 30.3846111)
   Min dTM/dt [1/us]::-5.08e+02 at an offset of 90.625 usec -> 2017-03-02T21:13:26.384.437.375 (XO: 30.3845787)
