-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170302_053437_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.073499 to 41.677348
-each separately from 3s to 41s and from 73499us to 677348us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.489988 in 1 packets - #/No: 18/18
      19.112994 in 1 packets - #/No: 36/18
MIU packet time from 2017-03-02T05:34:43.073.499.000 to 2017-03-02T05:35:21.677.348.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.489988 in 1 packets - #/No: 18/18
      19.112994 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4251456 to 42.4250109
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.9999327 in 1 packets - #/No: 18/18
      18.9999326 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170302_053437_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.4986454 to 4.4989344
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4899190  -- OK for SL --
   MIU time from 2017-03-02T05:34:43.073.499.000 to 2017-03-02T05:34:43.073.788.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.489988

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.9888534 to 23.9891424
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1129270  -- OK for SL --
   MIU time from 2017-03-02T05:35:02.563.776.000 to 2017-03-02T05:35:02.564.065.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.112994

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.1020694 to 43.1023584
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-03-02T05:35:21.677.059.000 to 2017-03-02T05:35:21.677.348.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-03-02T05:34:43.073.358.750
   Max SL corrected packet time: 2017-03-02T05:35:21.677.207.750
   Min SL corrected XO rel time: 3.0735001
   Max SL corrected XO rel time: 41.6773499
   TM values from -31 to 22 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-03-02T05:34:43.073.358.750 (XO: 3.0735001, t=34.0°C)  Rank: 00006
   TM values: Max =    22  Min =   -31  Ave =    -4.51  Std =     7.58
   Min TM value:-31 at an offset of 111.025 usec -> 2017-03-02T05:34:43.073.469.775 (XO: 3.0736111)
   Max TM value:22 at an offset of 160.251 usec -> 2017-03-02T05:34:43.073.519.000 (XO: 3.0736603)
   Max dTM/dt [1/us]::2.94e+02 at an offset of 118.888 usec -> 2017-03-02T05:34:43.073.477.637 (XO: 3.0736189)
   Min dTM/dt [1/us]::-3.21e+02 at an offset of 125.638 usec -> 2017-03-02T05:34:43.073.484.387 (XO: 3.0736257)

 - Block number: 2 starting at 2017-03-02T05:35:02.563.635.750 (XO: 22.5637771, t=34.0°C)  Rank: 00006
   TM values: Max =    22  Min =   -31  Ave =    -4.50  Std =     9.14
   Min TM value:-31 at an offset of 115.275 usec -> 2017-03-02T05:35:02.563.751.025 (XO: 22.5638923)
   Max TM value:22 at an offset of 97.900 usec -> 2017-03-02T05:35:02.563.733.650 (XO: 22.5638750)
   Max dTM/dt [1/us]::2.67e+02 at an offset of 114.163 usec -> 2017-03-02T05:35:02.563.749.912 (XO: 22.5638912)
   Min dTM/dt [1/us]::-2.72e+02 at an offset of 113.913 usec -> 2017-03-02T05:35:02.563.749.662 (XO: 22.5638910)

 - Block number: 3 starting at 2017-03-02T05:35:21.676.918.750 (XO: 41.6770609, t=34.0°C)  Rank: 00006
   TM values: Max =    15  Min =   -24  Ave =    -4.37  Std =     5.99
   Min TM value:-24 at an offset of 116.000 usec -> 2017-03-02T05:35:21.677.034.750 (XO: 41.6771769)
   Max TM value:15 at an offset of 109.388 usec -> 2017-03-02T05:35:21.677.028.137 (XO: 41.6771703)
   Max dTM/dt [1/us]::2.32e+02 at an offset of 102.850 usec -> 2017-03-02T05:35:21.677.021.600 (XO: 41.6771637)
   Min dTM/dt [1/us]::-2.35e+02 at an offset of 120.350 usec -> 2017-03-02T05:35:21.677.039.100 (XO: 41.6771812)
