-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170222_105544_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.269942 to 36.319890
-each separately from 0s to 36s and from 269942us to 668635us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.398115 in 1 packets - #/No: 18/18
      20.650966 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-22T10:55:47.269.942.000 to 2017-02-22T10:56:23.319.890.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.398115 in 1 packets - #/No: 18/18
      20.650966 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4259418 to 37.4258186
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999487 in 1 packets - #/No: 18/18
      20.9999281 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170222_105544_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.6958844 to 1.6961734
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3980630  -- OK for SL --
   MIU time from 2017-02-22T10:55:47.269.942.000 to 2017-02-22T10:55:47.270.231.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.398115

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0942364 to 17.0945254
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.6508950  -- OK for SL --
   MIU time from 2017-02-22T10:56:02.668.346.000 to 2017-02-22T10:56:02.668.635.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.650966

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.7454204 to 37.7457094
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-22T10:56:23.319.601.000 to 2017-02-22T10:56:23.319.890.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-22T10:55:47.269.801.750
   Max SL corrected packet time: 2017-02-22T10:56:23.319.749.750
   Min SL corrected XO rel time: 0.2699435
   Max SL corrected XO rel time: 36.3198919
   TM values from -37 to 28 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-22T10:55:47.269.801.750 (XO: 0.2699435, t=35.6°C)  Rank: 00006
   TM values: Max =    22  Min =   -34  Ave =    -4.58  Std =     7.70
   Min TM value:-34 at an offset of 124.713 usec -> 2017-02-22T10:55:47.269.926.462 (XO: 0.2700682)
   Max TM value:22 at an offset of 123.950 usec -> 2017-02-22T10:55:47.269.925.700 (XO: 0.2700675)
   Max dTM/dt [1/us]::2.16e+02 at an offset of 137.388 usec -> 2017-02-22T10:55:47.269.939.137 (XO: 0.2700809)
   Min dTM/dt [1/us]::-2.23e+02 at an offset of 188.188 usec -> 2017-02-22T10:55:47.269.989.938 (XO: 0.2701317)

 - Block number: 2 starting at 2017-02-22T10:56:02.668.205.750 (XO: 15.6683482, t=35.6°C)  Rank: 00005
   TM values: Max =    28  Min =   -37  Ave =    -4.45  Std =     9.20
   Min TM value:-37 at an offset of 90.000 usec -> 2017-02-22T10:56:02.668.295.750 (XO: 15.6684382)
   Max TM value:28 at an offset of 86.100 usec -> 2017-02-22T10:56:02.668.291.850 (XO: 15.6684343)
   Max dTM/dt [1/us]::2.72e+02 at an offset of 149.576 usec -> 2017-02-22T10:56:02.668.355.325 (XO: 15.6684978)
   Min dTM/dt [1/us]::-2.37e+02 at an offset of 100.213 usec -> 2017-02-22T10:56:02.668.305.962 (XO: 15.6684485)

 - Block number: 3 starting at 2017-02-22T10:56:23.319.460.750 (XO: 36.3196029, t=35.6°C)  Rank: 00005
   TM values: Max =    24  Min =   -29  Ave =    -4.66  Std =     7.42
   Min TM value:-29 at an offset of 112.300 usec -> 2017-02-22T10:56:23.319.573.050 (XO: 36.3197152)
   Max TM value:24 at an offset of 143.338 usec -> 2017-02-22T10:56:23.319.604.087 (XO: 36.3197462)
   Max dTM/dt [1/us]::1.87e+02 at an offset of 142.900 usec -> 2017-02-22T10:56:23.319.603.650 (XO: 36.3197458)
   Min dTM/dt [1/us]::-1.97e+02 at an offset of 296.089 usec -> 2017-02-22T10:56:23.319.756.838 (XO: 36.3198990)
