-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170222_095512_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.507496 to 40.288311
-each separately from 8s to 40s and from 85487us to 507785us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.577702 in 1 packets - #/No: 18/18
      25.202246 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-22T09:55:23.507.496.000 to 2017-02-22T09:55:55.288.311.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.577702 in 1 packets - #/No: 18/18
      25.202246 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4242525 to 41.4241419
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999758 in 1 packets - #/No: 18/18
      24.9999136 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170222_095512_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.9317479 to 9.9320369
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.5776790  -- OK for SL --
   MIU time from 2017-02-22T09:55:23.507.496.000 to 2017-02-22T09:55:23.507.785.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.577702

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5097159 to 16.5100049
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.2021590  -- OK for SL --
   MIU time from 2017-02-22T09:55:30.085.487.000 to 2017-02-22T09:55:30.085.776.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.202246

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.7121639 to 41.7124529
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-22T09:55:55.288.022.000 to 2017-02-22T09:55:55.288.311.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-22T09:55:23.507.355.750
   Max SL corrected packet time: 2017-02-22T09:55:55.288.170.750
   Min SL corrected XO rel time: 8.5074972
   Max SL corrected XO rel time: 40.2883120
   TM values from -44 to 33 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-22T09:55:23.507.355.750 (XO: 8.5074972, t=35.6°C)  Rank: 00008
   TM values: Max =    33  Min =   -44  Ave =    -4.50  Std =    10.59
   Min TM value:-44 at an offset of 159.626 usec -> 2017-02-22T09:55:23.507.515.375 (XO: 8.5076568)
   Max TM value:33 at an offset of 155.838 usec -> 2017-02-22T09:55:23.507.511.588 (XO: 8.5076530)
   Max dTM/dt [1/us]::4.65e+02 at an offset of 129.738 usec -> 2017-02-22T09:55:23.507.485.487 (XO: 8.5076269)
   Min dTM/dt [1/us]::-4.81e+02 at an offset of 175.876 usec -> 2017-02-22T09:55:23.507.531.625 (XO: 8.5076731)

 - Block number: 2 starting at 2017-02-22T09:55:30.085.346.750 (XO: 15.0854879, t=35.6°C)  Rank: 00008
   TM values: Max =    29  Min =   -41  Ave =    -4.44  Std =    10.41
   Min TM value:-41 at an offset of 160.001 usec -> 2017-02-22T09:55:30.085.506.750 (XO: 15.0856479)
   Max TM value:29 at an offset of 175.801 usec -> 2017-02-22T09:55:30.085.522.550 (XO: 15.0856637)
   Max dTM/dt [1/us]::4.11e+02 at an offset of 160.038 usec -> 2017-02-22T09:55:30.085.506.788 (XO: 15.0856479)
   Min dTM/dt [1/us]::-4.47e+02 at an offset of 166.863 usec -> 2017-02-22T09:55:30.085.513.613 (XO: 15.0856548)

 - Block number: 3 starting at 2017-02-22T09:55:55.287.881.750 (XO: 40.2880230, t=35.6°C)  Rank: 00007
   TM values: Max =    29  Min =   -39  Ave =    -4.40  Std =     9.61
   Min TM value:-39 at an offset of 104.175 usec -> 2017-02-22T09:55:55.287.985.925 (XO: 40.2881272)
   Max TM value:29 at an offset of 103.363 usec -> 2017-02-22T09:55:55.287.985.112 (XO: 40.2881263)
   Max dTM/dt [1/us]::2.76e+02 at an offset of 282.651 usec -> 2017-02-22T09:55:55.288.164.400 (XO: 40.2883056)
   Min dTM/dt [1/us]::-3.03e+02 at an offset of 235.001 usec -> 2017-02-22T09:55:55.288.116.750 (XO: 40.2882580)
