-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170220_193556_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.467585 to 39.463719
-each separately from 6s to 39s and from 385393us to 467874us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.917519 in 1 packets - #/No: 18/18
      24.077748 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-20T19:36:05.467.585.000 to 2017-02-20T19:36:38.463.719.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.917519 in 1 packets - #/No: 18/18
      24.077748 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4309727 to 40.4308566
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999683 in 1 packets - #/No: 18/18
      23.9999156 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170220_193556_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8985581 to 7.8988471
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9174870  -- OK for SL --
   MIU time from 2017-02-20T19:36:05.467.585.000 to 2017-02-20T19:36:05.467.874.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.917519

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8163341 to 16.8166231
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.0776630  -- OK for SL --
   MIU time from 2017-02-20T19:36:14.385.393.000 to 2017-02-20T19:36:14.385.682.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.077748

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.8942861 to 40.8945751
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-20T19:36:38.463.430.000 to 2017-02-20T19:36:38.463.719.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-20T19:36:05.467.444.750
   Max SL corrected packet time: 2017-02-20T19:36:38.463.578.750
   Min SL corrected XO rel time: 6.4675870
   Max SL corrected XO rel time: 39.4637201
   TM values from -60 to 56 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-20T19:36:05.467.444.750 (XO: 6.4675870, t=35.6°C)  Rank: 00011
   TM values: Max =    55  Min =   -59  Ave =    -4.41  Std =    13.64
   Min TM value:-59 at an offset of 107.950 usec -> 2017-02-20T19:36:05.467.552.700 (XO: 6.4676950)
   Max TM value:55 at an offset of 53.963 usec -> 2017-02-20T19:36:05.467.498.712 (XO: 6.4676410)
   Max dTM/dt [1/us]::7.27e+02 at an offset of 106.500 usec -> 2017-02-20T19:36:05.467.551.250 (XO: 6.4676935)
   Min dTM/dt [1/us]::-7.59e+02 at an offset of 137.413 usec -> 2017-02-20T19:36:05.467.582.162 (XO: 6.4677244)

 - Block number: 2 starting at 2017-02-20T19:36:14.385.252.750 (XO: 15.3853944, t=35.6°C)  Rank: 00009
   TM values: Max =    45  Min =   -49  Ave =    -4.49  Std =    12.08
   Min TM value:-49 at an offset of 85.650 usec -> 2017-02-20T19:36:14.385.338.400 (XO: 15.3854801)
   Max TM value:45 at an offset of 86.550 usec -> 2017-02-20T19:36:14.385.339.300 (XO: 15.3854810)
   Max dTM/dt [1/us]::6.31e+02 at an offset of 293.551 usec -> 2017-02-20T19:36:14.385.546.301 (XO: 15.3856880)
   Min dTM/dt [1/us]::-6.78e+02 at an offset of 117.150 usec -> 2017-02-20T19:36:14.385.369.900 (XO: 15.3855116)

 - Block number: 3 starting at 2017-02-20T19:36:38.463.289.750 (XO: 39.4634311, t=35.6°C)  Rank: 00011
   TM values: Max =    56  Min =   -60  Ave =    -4.51  Std =    14.32
   Min TM value:-60 at an offset of 95.538 usec -> 2017-02-20T19:36:38.463.385.287 (XO: 39.4635267)
   Max TM value:56 at an offset of 84.738 usec -> 2017-02-20T19:36:38.463.374.487 (XO: 39.4635159)
   Max dTM/dt [1/us]::8.89e+02 at an offset of 95.563 usec -> 2017-02-20T19:36:38.463.385.312 (XO: 39.4635267)
   Min dTM/dt [1/us]::-8.85e+02 at an offset of 94.675 usec -> 2017-02-20T19:36:38.463.384.425 (XO: 39.4635258)
