-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170220_155142_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.443815 to 41.172719
-each separately from 7s to 41s and from 159987us to 444104us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.715883 in 1 packets - #/No: 18/18
      26.012154 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-20T15:51:52.443.815.000 to 2017-02-20T15:52:26.172.719.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.715883 in 1 packets - #/No: 18/18
      26.012154 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4304273 to 42.4303075
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999717 in 1 packets - #/No: 18/18
      25.9999085 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170220_155142_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.8742425 to 8.8745315
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7158550  -- OK for SL --
   MIU time from 2017-02-20T15:51:52.443.815.000 to 2017-02-20T15:51:52.444.104.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.715883

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5903865 to 16.5906755
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.0120630  -- OK for SL --
   MIU time from 2017-02-20T15:52:00.159.987.000 to 2017-02-20T15:52:00.160.276.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.012154

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.6027385 to 42.6030275
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-20T15:52:26.172.430.000 to 2017-02-20T15:52:26.172.719.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-20T15:51:52.443.674.750
   Max SL corrected packet time: 2017-02-20T15:52:26.172.578.750
   Min SL corrected XO rel time: 7.4438168
   Max SL corrected XO rel time: 41.1727206
   TM values from -49 to 43 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-20T15:51:52.443.674.750 (XO: 7.4438168, t=35.6°C)  Rank: 00009
   TM values: Max =    37  Min =   -49  Ave =    -4.47  Std =    12.62
   Min TM value:-49 at an offset of 165.813 usec -> 2017-02-20T15:51:52.443.840.563 (XO: 7.4439826)
   Max TM value:37 at an offset of 140.751 usec -> 2017-02-20T15:51:52.443.815.500 (XO: 7.4439575)
   Max dTM/dt [1/us]::4.66e+02 at an offset of 183.176 usec -> 2017-02-20T15:51:52.443.857.925 (XO: 7.4440000)
   Min dTM/dt [1/us]::-4.87e+02 at an offset of 182.413 usec -> 2017-02-20T15:51:52.443.857.163 (XO: 7.4439992)

 - Block number: 2 starting at 2017-02-20T15:52:00.159.846.750 (XO: 15.1599881, t=35.6°C)  Rank: 00011
   TM values: Max =    43  Min =   -49  Ave =    -4.48  Std =    13.87
   Min TM value:-49 at an offset of 239.688 usec -> 2017-02-20T15:52:00.160.086.438 (XO: 15.1602277)
   Max TM value:43 at an offset of 178.363 usec -> 2017-02-20T15:52:00.160.025.113 (XO: 15.1601664)
   Max dTM/dt [1/us]::4.75e+02 at an offset of 170.588 usec -> 2017-02-20T15:52:00.160.017.338 (XO: 15.1601586)
   Min dTM/dt [1/us]::-5.04e+02 at an offset of 241.026 usec -> 2017-02-20T15:52:00.160.087.775 (XO: 15.1602291)

 - Block number: 3 starting at 2017-02-20T15:52:26.172.289.750 (XO: 41.1724316, t=35.6°C)  Rank: 00010
   TM values: Max =    33  Min =   -45  Ave =    -4.46  Std =    11.08
   Min TM value:-45 at an offset of 289.276 usec -> 2017-02-20T15:52:26.172.579.026 (XO: 41.1727209)
   Max TM value:33 at an offset of 288.901 usec -> 2017-02-20T15:52:26.172.578.651 (XO: 41.1727205)
   Max dTM/dt [1/us]::4.04e+02 at an offset of 288.876 usec -> 2017-02-20T15:52:26.172.578.626 (XO: 41.1727205)
   Min dTM/dt [1/us]::-4.61e+02 at an offset of 288.926 usec -> 2017-02-20T15:52:26.172.578.676 (XO: 41.1727205)
