-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170216_094346_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.105501 to 44.945188
-each separately from 0s to 44s and from 105501us to 945188us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.687673 in 1 packets - #/No: 18/18
      27.151147 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-16T09:43:49.105.501.000 to 2017-02-16T09:44:33.945.188.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.687673 in 1 packets - #/No: 18/18
      27.151147 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4302303 to 45.4300819
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999427 in 1 packets - #/No: 18/18
      26.9999089 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170216_094346_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5357317 to 1.5360207
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6876150  -- OK for SL --
   MIU time from 2017-02-16T09:43:49.105.501.000 to 2017-02-16T09:43:49.105.790.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.687673

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.2236357 to 19.2239247
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.1510550  -- OK for SL --
   MIU time from 2017-02-16T09:44:06.793.463.000 to 2017-02-16T09:44:06.793.752.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.151147

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.3749797 to 46.3752687
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-16T09:44:33.944.899.000 to 2017-02-16T09:44:33.945.188.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-16T09:43:49.105.360.750
   Max SL corrected packet time: 2017-02-16T09:44:33.945.047.750
   Min SL corrected XO rel time: 0.1055018
   Max SL corrected XO rel time: 44.9451912
   TM values from -31 to 20 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-16T09:43:49.105.360.750 (XO: 0.1055018, t=35.6°C)  Rank: 00004
   TM values: Max =    20  Min =   -31  Ave =    -4.55  Std =     8.04
   Min TM value:-31 at an offset of 131.150 usec -> 2017-02-16T09:43:49.105.491.900 (XO: 0.1056329)
   Max TM value:20 at an offset of 111.963 usec -> 2017-02-16T09:43:49.105.472.712 (XO: 0.1056137)
   Max dTM/dt [1/us]::2.25e+02 at an offset of 146.838 usec -> 2017-02-16T09:43:49.105.507.587 (XO: 0.1056486)
   Min dTM/dt [1/us]::-2.28e+02 at an offset of 180.938 usec -> 2017-02-16T09:43:49.105.541.688 (XO: 0.1056827)

 - Block number: 2 starting at 2017-02-16T09:44:06.793.322.750 (XO: 17.7934654, t=35.6°C)  Rank: 00005
   TM values: Max =    17  Min =   -27  Ave =    -4.47  Std =     5.87
   Min TM value:-27 at an offset of 76.163 usec -> 2017-02-16T09:44:06.793.398.912 (XO: 17.7935416)
   Max TM value:17 at an offset of 121.013 usec -> 2017-02-16T09:44:06.793.443.762 (XO: 17.7935864)
   Max dTM/dt [1/us]::1.68e+02 at an offset of 111.388 usec -> 2017-02-16T09:44:06.793.434.137 (XO: 17.7935768)
   Min dTM/dt [1/us]::-2.19e+02 at an offset of 112.263 usec -> 2017-02-16T09:44:06.793.435.012 (XO: 17.7935777)

 - Block number: 3 starting at 2017-02-16T09:44:33.944.758.750 (XO: 44.9449022, t=35.6°C)  Rank: 00005
   TM values: Max =    18  Min =   -29  Ave =    -4.52  Std =     7.02
   Min TM value:-29 at an offset of 133.688 usec -> 2017-02-16T09:44:33.944.892.437 (XO: 44.9450359)
   Max TM value:18 at an offset of 132.875 usec -> 2017-02-16T09:44:33.944.891.625 (XO: 44.9450351)
   Max dTM/dt [1/us]::1.64e+02 at an offset of 247.738 usec -> 2017-02-16T09:44:33.945.006.488 (XO: 44.9451500)
   Min dTM/dt [1/us]::-1.66e+02 at an offset of 218.851 usec -> 2017-02-16T09:44:33.944.977.600 (XO: 44.9451211)
