-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170216_084314_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.875746 to 33.826290
-each separately from 1s to 33s and from 537226us to 876035us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.661191 in 1 packets - #/No: 18/18
      11.288486 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-16T08:43:18.875.746.000 to 2017-02-16T08:43:50.826.290.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.661191 in 1 packets - #/No: 18/18
      11.288486 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4274175 to 34.4273069
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999274 in 1 packets - #/No: 18/18
      10.9999620 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170216_084314_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3031627 to 3.3034517
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.6611190  -- OK for SL --
   MIU time from 2017-02-16T08:43:18.875.746.000 to 2017-02-16T08:43:18.876.035.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.661191

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.9645707 to 23.9648597
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.2884470  -- OK for SL --
   MIU time from 2017-02-16T08:43:39.537.226.000 to 2017-02-16T08:43:39.537.515.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.288486

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.2533067 to 35.2535957
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-16T08:43:50.826.001.000 to 2017-02-16T08:43:50.826.290.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-16T08:43:18.875.605.750
   Max SL corrected packet time: 2017-02-16T08:43:50.826.149.750
   Min SL corrected XO rel time: 1.8757483
   Max SL corrected XO rel time: 33.8262917
   TM values from -47 to 39 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-16T08:43:18.875.605.750 (XO: 1.8757483, t=35.6°C)  Rank: 00010
   TM values: Max =    36  Min =   -45  Ave =    -4.52  Std =    11.95
   Min TM value:-45 at an offset of 56.125 usec -> 2017-02-16T08:43:18.875.661.875 (XO: 1.8758044)
   Max TM value:36 at an offset of 157.401 usec -> 2017-02-16T08:43:18.875.763.150 (XO: 1.8759057)
   Max dTM/dt [1/us]::2.49e+02 at an offset of 123.013 usec -> 2017-02-16T08:43:18.875.728.762 (XO: 1.8758713)
   Min dTM/dt [1/us]::-2.32e+02 at an offset of 68.838 usec -> 2017-02-16T08:43:18.875.674.587 (XO: 1.8758171)

 - Block number: 2 starting at 2017-02-16T08:43:39.537.085.750 (XO: 22.5372277, t=35.6°C)  Rank: 00011
   TM values: Max =    39  Min =   -46  Ave =    -4.54  Std =    13.36
   Min TM value:-46 at an offset of 126.425 usec -> 2017-02-16T08:43:39.537.212.175 (XO: 22.5373541)
   Max TM value:39 at an offset of 155.913 usec -> 2017-02-16T08:43:39.537.241.663 (XO: 22.5373836)
   Max dTM/dt [1/us]::3.13e+02 at an offset of 193.901 usec -> 2017-02-16T08:43:39.537.279.650 (XO: 22.5374216)
   Min dTM/dt [1/us]::-2.96e+02 at an offset of 87.625 usec -> 2017-02-16T08:43:39.537.173.375 (XO: 22.5373153)

 - Block number: 3 starting at 2017-02-16T08:43:50.825.860.750 (XO: 33.8260027, t=35.6°C)  Rank: 00011
   TM values: Max =    38  Min =   -47  Ave =    -4.75  Std =    13.00
   Min TM value:-47 at an offset of 132.275 usec -> 2017-02-16T08:43:50.825.993.025 (XO: 33.8261350)
   Max TM value:38 at an offset of 146.676 usec -> 2017-02-16T08:43:50.826.007.425 (XO: 33.8261494)
   Max dTM/dt [1/us]::2.70e+02 at an offset of 153.001 usec -> 2017-02-16T08:43:50.826.013.750 (XO: 33.8261557)
   Min dTM/dt [1/us]::-2.88e+02 at an offset of 151.276 usec -> 2017-02-16T08:43:50.826.012.025 (XO: 33.8261540)
