-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170214_202402_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.599218 to 31.728392
-each separately from 1s to 31s and from 78515us to 728392us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.479008 in 1 packets - #/No: 18/18
      15.649299 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-14T20:24:06.599.218.000 to 2017-02-14T20:24:36.728.392.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.479008 in 1 packets - #/No: 18/18
      15.649299 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4218994 to 32.4217981
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999494 in 1 packets - #/No: 18/18
      14.9999493 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170214_202402_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.0211169 to 3.0214059
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4789590  -- OK for SL --
   MIU time from 2017-02-14T20:24:06.599.218.000 to 2017-02-14T20:24:06.599.507.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.479008

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.5003649 to 17.5006539
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6492470  -- OK for SL --
   MIU time from 2017-02-14T20:24:21.078.515.000 to 2017-02-14T20:24:21.078.804.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.649299

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.1499009 to 33.1501899
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-14T20:24:36.728.103.000 to 2017-02-14T20:24:36.728.392.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-14T20:24:06.599.077.750
   Max SL corrected packet time: 2017-02-14T20:24:36.728.251.750
   Min SL corrected XO rel time: 1.5992195
   Max SL corrected XO rel time: 31.7283942
   TM values from -38 to 30 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-14T20:24:06.599.077.750 (XO: 1.5992195, t=35.6°C)  Rank: 00007
   TM values: Max =    30  Min =   -38  Ave =    -4.49  Std =     7.95
   Min TM value:-38 at an offset of 123.788 usec -> 2017-02-14T20:24:06.599.201.537 (XO: 1.5993433)
   Max TM value:30 at an offset of 123.313 usec -> 2017-02-14T20:24:06.599.201.062 (XO: 1.5993428)
   Max dTM/dt [1/us]::4.60e+02 at an offset of 121.513 usec -> 2017-02-14T20:24:06.599.199.262 (XO: 1.5993410)
   Min dTM/dt [1/us]::-4.47e+02 at an offset of 121.575 usec -> 2017-02-14T20:24:06.599.199.325 (XO: 1.5993411)

 - Block number: 2 starting at 2017-02-14T20:24:21.078.374.750 (XO: 16.0785164, t=35.6°C)  Rank: 00007
   TM values: Max =    25  Min =   -38  Ave =    -4.56  Std =     8.38
   Min TM value:-38 at an offset of 131.425 usec -> 2017-02-14T20:24:21.078.506.175 (XO: 16.0786478)
   Max TM value:25 at an offset of 165.751 usec -> 2017-02-14T20:24:21.078.540.500 (XO: 16.0786821)
   Max dTM/dt [1/us]::3.73e+02 at an offset of 131.463 usec -> 2017-02-14T20:24:21.078.506.212 (XO: 16.0786478)
   Min dTM/dt [1/us]::-3.77e+02 at an offset of 131.388 usec -> 2017-02-14T20:24:21.078.506.137 (XO: 16.0786478)

 - Block number: 3 starting at 2017-02-14T20:24:36.727.962.750 (XO: 31.7281052, t=35.6°C)  Rank: 00008
   TM values: Max =    25  Min =   -38  Ave =    -4.47  Std =     7.87
   Min TM value:-38 at an offset of 294.413 usec -> 2017-02-14T20:24:36.728.257.163 (XO: 31.7283996)
   Max TM value:25 at an offset of 88.650 usec -> 2017-02-14T20:24:36.728.051.400 (XO: 31.7281939)
   Max dTM/dt [1/us]::4.61e+02 at an offset of 168.076 usec -> 2017-02-14T20:24:36.728.130.825 (XO: 31.7282733)
   Min dTM/dt [1/us]::-4.37e+02 at an offset of 95.050 usec -> 2017-02-14T20:24:36.728.057.800 (XO: 31.7282003)
