-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170210_230125_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.642157 to 44.537138
-each separately from 10s to 44s and from 437275us to 642446us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.794829 in 1 packets - #/No: 18/18
      25.099285 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-10T23:01:38.642.157.000 to 2017-02-10T23:02:12.537.138.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.794829 in 1 packets - #/No: 18/18
      25.099285 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4301287 to 45.4300130
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999693 in 1 packets - #/No: 18/18
      24.9999150 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170210_230125_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.0722854 to 12.0725744
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7947990  -- OK for SL --
   MIU time from 2017-02-10T23:01:38.642.157.000 to 2017-02-10T23:01:38.642.446.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.794829

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.8673734 to 20.8676624
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.0991990  -- OK for SL --
   MIU time from 2017-02-10T23:01:47.437.275.000 to 2017-02-10T23:01:47.437.564.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.099285

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.9668614 to 45.9671504
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-10T23:02:12.536.849.000 to 2017-02-10T23:02:12.537.138.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-10T23:01:38.642.016.750
   Max SL corrected packet time: 2017-02-10T23:02:12.536.997.750
   Min SL corrected XO rel time: 10.6421589
   Max SL corrected XO rel time: 44.5371392
   TM values from -33 to 21 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-10T23:01:38.642.016.750 (XO: 10.6421589, t=35.6°C)  Rank: 00006
   TM values: Max =    21  Min =   -28  Ave =    -4.51  Std =     6.65
   Min TM value:-28 at an offset of 122.013 usec -> 2017-02-10T23:01:38.642.138.762 (XO: 10.6422810)
   Max TM value:21 at an offset of 201.963 usec -> 2017-02-10T23:01:38.642.218.713 (XO: 10.6423609)
   Max dTM/dt [1/us]::2.44e+02 at an offset of 99.463 usec -> 2017-02-10T23:01:38.642.116.212 (XO: 10.6422584)
   Min dTM/dt [1/us]::-2.49e+02 at an offset of 124.763 usec -> 2017-02-10T23:01:38.642.141.512 (XO: 10.6422837)

 - Block number: 2 starting at 2017-02-10T23:01:47.437.134.750 (XO: 19.4372769, t=35.6°C)  Rank: 00006
   TM values: Max =    20  Min =   -33  Ave =    -4.53  Std =     6.67
   Min TM value:-33 at an offset of 116.700 usec -> 2017-02-10T23:01:47.437.251.450 (XO: 19.4373936)
   Max TM value:20 at an offset of 79.388 usec -> 2017-02-10T23:01:47.437.214.137 (XO: 19.4373563)
   Max dTM/dt [1/us]::3.08e+02 at an offset of 234.151 usec -> 2017-02-10T23:01:47.437.368.900 (XO: 19.4375110)
   Min dTM/dt [1/us]::-2.75e+02 at an offset of 75.988 usec -> 2017-02-10T23:01:47.437.210.737 (XO: 19.4373529)

 - Block number: 3 starting at 2017-02-10T23:02:12.536.708.750 (XO: 44.5368502, t=35.6°C)  Rank: 00006
   TM values: Max =    21  Min =   -28  Ave =    -4.57  Std =     6.89
   Min TM value:-28 at an offset of 94.438 usec -> 2017-02-10T23:02:12.536.803.187 (XO: 44.5369447)
   Max TM value:21 at an offset of 111.775 usec -> 2017-02-10T23:02:12.536.820.525 (XO: 44.5369620)
   Max dTM/dt [1/us]::3.31e+02 at an offset of 51.875 usec -> 2017-02-10T23:02:12.536.760.625 (XO: 44.5369021)
   Min dTM/dt [1/us]::-2.90e+02 at an offset of 292.901 usec -> 2017-02-10T23:02:12.537.001.650 (XO: 44.5371431)
