-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170209_234825_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.918725 to 32.917167
-each separately from 11s to 32s and from 499212us to 919014us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.580198 in 1 packets - #/No: 18/18
      14.417377 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-09T23:48:39.918.725.000 to 2017-02-09T23:49:00.917.167.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.580198 in 1 packets - #/No: 18/18
      14.417377 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.4253154 to 33.4252433
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999759 in 1 packets - #/No: 18/18
      13.9999520 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170209_234825_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.3440394 to 13.3443284
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.5801750  -- OK for SL --
   MIU time from 2017-02-09T23:48:39.918.725.000 to 2017-02-09T23:48:39.919.014.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.580198

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.9245034 to 19.9247924
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4173270  -- OK for SL --
   MIU time from 2017-02-09T23:48:46.499.212.000 to 2017-02-09T23:48:46.499.501.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.417377

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.3421194 to 34.3424084
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-09T23:49:00.916.878.000 to 2017-02-09T23:49:00.917.167.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-09T23:48:39.918.584.750
   Max SL corrected packet time: 2017-02-09T23:49:00.917.026.750
   Min SL corrected XO rel time: 11.9187272
   Max SL corrected XO rel time: 32.9171683
   TM values from -47 to 42 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-09T23:48:39.918.584.750 (XO: 11.9187272, t=34.0°C)  Rank: 00005
   TM values: Max =    22  Min =   -33  Ave =    -4.63  Std =     7.70
   Min TM value:-33 at an offset of 197.501 usec -> 2017-02-09T23:48:39.918.782.250 (XO: 11.9189247)
   Max TM value:22 at an offset of 141.375 usec -> 2017-02-09T23:48:39.918.726.125 (XO: 11.9188686)
   Max dTM/dt [1/us]::3.69e+02 at an offset of 142.713 usec -> 2017-02-09T23:48:39.918.727.462 (XO: 11.9188699)
   Min dTM/dt [1/us]::-3.69e+02 at an offset of 180.738 usec -> 2017-02-09T23:48:39.918.765.488 (XO: 11.9189080)

 - Block number: 2 starting at 2017-02-09T23:48:46.499.071.750 (XO: 18.4992138, t=34.0°C)  Rank: 00008
   TM values: Max =    42  Min =   -47  Ave =    -4.51  Std =     9.44
   Min TM value:-47 at an offset of 114.725 usec -> 2017-02-09T23:48:46.499.186.475 (XO: 18.4993285)
   Max TM value:42 at an offset of 119.363 usec -> 2017-02-09T23:48:46.499.191.112 (XO: 18.4993332)
   Max dTM/dt [1/us]::6.65e+02 at an offset of 119.850 usec -> 2017-02-09T23:48:46.499.191.600 (XO: 18.4993336)
   Min dTM/dt [1/us]::-5.48e+02 at an offset of 131.738 usec -> 2017-02-09T23:48:46.499.203.487 (XO: 18.4993455)

 - Block number: 3 starting at 2017-02-09T23:49:00.916.737.750 (XO: 32.9168793, t=34.0°C)  Rank: 00007
   TM values: Max =    26  Min =   -36  Ave =    -4.56  Std =     9.28
   Min TM value:-36 at an offset of 122.275 usec -> 2017-02-09T23:49:00.916.860.025 (XO: 32.9170016)
   Max TM value:26 at an offset of 204.513 usec -> 2017-02-09T23:49:00.916.942.263 (XO: 32.9170838)
   Max dTM/dt [1/us]::3.68e+02 at an offset of 83.675 usec -> 2017-02-09T23:49:00.916.821.425 (XO: 32.9169630)
   Min dTM/dt [1/us]::-4.14e+02 at an offset of 122.238 usec -> 2017-02-09T23:49:00.916.859.987 (XO: 32.9170015)
