-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170208_165408_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.487960 to 31.789058
-each separately from 0s to 31s and from 487960us to 811805us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.323267 in 1 packets - #/No: 18/18
      10.976964 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-08T16:54:11.487.960.000 to 2017-02-08T16:54:42.789.058.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.323267 in 1 packets - #/No: 18/18
      10.976964 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4289429 to 32.4288381
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.9999324 in 1 packets - #/No: 18/18
      10.9999628 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170208_165408_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9169025 to 1.9171915
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.3231990  -- OK for SL --
   MIU time from 2017-02-08T16:54:11.487.960.000 to 2017-02-08T16:54:11.488.249.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.323267

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.2403905 to 22.2406795
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9769270  -- OK for SL --
   MIU time from 2017-02-08T16:54:31.811.516.000 to 2017-02-08T16:54:31.811.805.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.976964

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.2176065 to 33.2178955
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-08T16:54:42.788.769.000 to 2017-02-08T16:54:42.789.058.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-08T16:54:11.487.819.750
   Max SL corrected packet time: 2017-02-08T16:54:42.788.917.750
   Min SL corrected XO rel time: 0.4879613
   Max SL corrected XO rel time: 31.7890601
   TM values from -46 to 31 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-08T16:54:11.487.819.750 (XO: 0.4879613, t=35.6°C)  Rank: 00008
   TM values: Max =    29  Min =   -46  Ave =    -4.57  Std =    10.75
   Min TM value:-46 at an offset of 67.450 usec -> 2017-02-08T16:54:11.487.887.200 (XO: 0.4880287)
   Max TM value:29 at an offset of 57.475 usec -> 2017-02-08T16:54:11.487.877.225 (XO: 0.4880187)
   Max dTM/dt [1/us]::5.62e+02 at an offset of 67.500 usec -> 2017-02-08T16:54:11.487.887.250 (XO: 0.4880288)
   Min dTM/dt [1/us]::-5.14e+02 at an offset of 136.938 usec -> 2017-02-08T16:54:11.487.956.687 (XO: 0.4880982)

 - Block number: 2 starting at 2017-02-08T16:54:31.811.375.750 (XO: 20.8115180, t=35.6°C)  Rank: 00009
   TM values: Max =    31  Min =   -40  Ave =    -4.59  Std =     9.52
   Min TM value:-40 at an offset of 128.288 usec -> 2017-02-08T16:54:31.811.504.037 (XO: 20.8116462)
   Max TM value:31 at an offset of 126.700 usec -> 2017-02-08T16:54:31.811.502.450 (XO: 20.8116447)
   Max dTM/dt [1/us]::5.89e+02 at an offset of 126.663 usec -> 2017-02-08T16:54:31.811.502.412 (XO: 20.8116446)
   Min dTM/dt [1/us]::-5.50e+02 at an offset of 101.113 usec -> 2017-02-08T16:54:31.811.476.862 (XO: 20.8116191)

 - Block number: 3 starting at 2017-02-08T16:54:42.788.628.750 (XO: 31.7887711, t=35.6°C)  Rank: 00009
   TM values: Max =    31  Min =   -44  Ave =    -4.43  Std =    11.40
   Min TM value:-44 at an offset of 243.513 usec -> 2017-02-08T16:54:42.788.872.263 (XO: 31.7890146)
   Max TM value:31 at an offset of 170.238 usec -> 2017-02-08T16:54:42.788.798.988 (XO: 31.7889413)
   Max dTM/dt [1/us]::5.64e+02 at an offset of 125.300 usec -> 2017-02-08T16:54:42.788.754.050 (XO: 31.7888964)
   Min dTM/dt [1/us]::-5.48e+02 at an offset of 124.713 usec -> 2017-02-08T16:54:42.788.753.462 (XO: 31.7888958)
