-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170207_211303_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.922065 to 37.946667
-each separately from 1s to 37s and from 140398us to 946667us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.218044 in 1 packets - #/No: 18/18
      22.805691 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-07T21:13:07.922.065.000 to 2017-02-07T21:13:43.946.667.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.218044 in 1 packets - #/No: 18/18
      22.805691 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4251864 to 38.4250656
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999529 in 1 packets - #/No: 18/18
      21.9999263 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170207_211303_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3472500 to 3.3475390
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2179990  -- OK for SL --
   MIU time from 2017-02-07T21:13:07.922.065.000 to 2017-02-07T21:13:07.922.354.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.218044

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5655380 to 16.5658270
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.8056150  -- OK for SL --
   MIU time from 2017-02-07T21:13:21.140.398.000 to 2017-02-07T21:13:21.140.687.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.805691

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.3714420 to 39.3717310
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-07T21:13:43.946.378.000 to 2017-02-07T21:13:43.946.667.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-07T21:13:07.921.924.750
   Max SL corrected packet time: 2017-02-07T21:13:43.946.526.750
   Min SL corrected XO rel time: 1.9220667
   Max SL corrected XO rel time: 37.9466685
   TM values from -46 to 33 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-07T21:13:07.921.924.750 (XO: 1.9220667, t=35.6°C)  Rank: 00009
   TM values: Max =    28  Min =   -34  Ave =    -4.53  Std =     8.49
   Min TM value:-34 at an offset of 185.351 usec -> 2017-02-07T21:13:07.922.110.100 (XO: 1.9222521)
   Max TM value:28 at an offset of 122.250 usec -> 2017-02-07T21:13:07.922.047.000 (XO: 1.9221890)
   Max dTM/dt [1/us]::3.27e+02 at an offset of 132.300 usec -> 2017-02-07T21:13:07.922.057.050 (XO: 1.9221990)
   Min dTM/dt [1/us]::-3.51e+02 at an offset of 205.013 usec -> 2017-02-07T21:13:07.922.129.763 (XO: 1.9222717)

 - Block number: 2 starting at 2017-02-07T21:13:21.140.257.750 (XO: 15.1403992, t=35.6°C)  Rank: 00009
   TM values: Max =    33  Min =   -46  Ave =    -4.53  Std =     9.23
   Min TM value:-46 at an offset of 133.200 usec -> 2017-02-07T21:13:21.140.390.950 (XO: 15.1405324)
   Max TM value:33 at an offset of 130.813 usec -> 2017-02-07T21:13:21.140.388.562 (XO: 15.1405300)
   Max dTM/dt [1/us]::3.65e+02 at an offset of 39.475 usec -> 2017-02-07T21:13:21.140.297.225 (XO: 15.1404386)
   Min dTM/dt [1/us]::-3.74e+02 at an offset of 141.763 usec -> 2017-02-07T21:13:21.140.399.512 (XO: 15.1405409)

 - Block number: 3 starting at 2017-02-07T21:13:43.946.237.750 (XO: 37.9463795, t=35.6°C)  Rank: 00008
   TM values: Max =    28  Min =   -44  Ave =    -4.57  Std =     9.22
   Min TM value:-44 at an offset of 39.813 usec -> 2017-02-07T21:13:43.946.277.562 (XO: 37.9464193)
   Max TM value:28 at an offset of 239.751 usec -> 2017-02-07T21:13:43.946.477.500 (XO: 37.9466193)
   Max dTM/dt [1/us]::4.15e+02 at an offset of 241.213 usec -> 2017-02-07T21:13:43.946.478.963 (XO: 37.9466207)
   Min dTM/dt [1/us]::-4.01e+02 at an offset of 239.788 usec -> 2017-02-07T21:13:43.946.477.538 (XO: 37.9466193)
