-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170207_113759_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.165798 to 31.434108
-each separately from 11s to 31s and from 165798us to 434108us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.003164 in 1 packets - #/No: 18/18
      16.264279 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-07T11:38:13.165.798.000 to 2017-02-07T11:38:33.434.108.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.003164 in 1 packets - #/No: 18/18
      16.264279 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9259387 to 32.9258701
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.9999863 in 1 packets - #/No: 18/18
      15.9999451 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      04 -> 49.8°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170207_113759_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.0917370 to 13.0920260
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.0031510  -- OK for SL --
   MIU time from 2017-02-07T11:38:13.165.798.000 to 2017-02-07T11:38:13.166.087.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.003164

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0951770 to 17.0954660
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.2642230  -- OK for SL --
   MIU time from 2017-02-07T11:38:17.169.251.000 to 2017-02-07T11:38:17.169.540.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.264279

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.3596890 to 33.3599780
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-07T11:38:33.433.819.000 to 2017-02-07T11:38:33.434.108.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-07T11:38:13.165.657.750
   Max SL corrected packet time: 2017-02-07T11:38:33.433.967.750
   Min SL corrected XO rel time: 11.1657989
   Max SL corrected XO rel time: 31.4341094
   TM values from -37 to 29 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-07T11:38:13.165.657.750 (XO: 11.1657989, t=35.6°C)  Rank: 00006
   TM values: Max =    29  Min =   -37  Ave =    -4.48  Std =     9.32
   Min TM value:-37 at an offset of 126.850 usec -> 2017-02-07T11:38:13.165.784.600 (XO: 11.1659257)
   Max TM value:29 at an offset of 63.625 usec -> 2017-02-07T11:38:13.165.721.375 (XO: 11.1658625)
   Max dTM/dt [1/us]::2.38e+02 at an offset of 63.575 usec -> 2017-02-07T11:38:13.165.721.325 (XO: 11.1658624)
   Min dTM/dt [1/us]::-2.53e+02 at an offset of 149.476 usec -> 2017-02-07T11:38:13.165.807.225 (XO: 11.1659483)

 - Block number: 2 starting at 2017-02-07T11:38:17.169.110.750 (XO: 15.1692526, t=35.6°C)  Rank: 00005
   TM values: Max =    21  Min =   -30  Ave =    -4.52  Std =     6.93
   Min TM value:-30 at an offset of 74.988 usec -> 2017-02-07T11:38:17.169.185.737 (XO: 15.1693276)
   Max TM value:21 at an offset of 104.450 usec -> 2017-02-07T11:38:17.169.215.200 (XO: 15.1693570)
   Max dTM/dt [1/us]::1.86e+02 at an offset of 104.463 usec -> 2017-02-07T11:38:17.169.215.212 (XO: 15.1693570)
   Min dTM/dt [1/us]::-2.17e+02 at an offset of 304.039 usec -> 2017-02-07T11:38:17.169.414.788 (XO: 15.1695566)

 - Block number: 3 starting at 2017-02-07T11:38:33.433.678.750 (XO: 31.4338204, t=35.6°C)  Rank: 00006
   TM values: Max =    24  Min =   -33  Ave =    -4.55  Std =     7.81
   Min TM value:-33 at an offset of 150.938 usec -> 2017-02-07T11:38:33.433.829.688 (XO: 31.4339713)
   Max TM value:24 at an offset of 121.213 usec -> 2017-02-07T11:38:33.433.799.962 (XO: 31.4339416)
   Max dTM/dt [1/us]::2.20e+02 at an offset of 151.376 usec -> 2017-02-07T11:38:33.433.830.125 (XO: 31.4339718)
   Min dTM/dt [1/us]::-2.05e+02 at an offset of 126.125 usec -> 2017-02-07T11:38:33.433.804.875 (XO: 31.4339465)
