-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170206_104853_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.573132 to 30.326642
-each separately from 10s to 30s and from 146156us to 573421us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.572735 in 1 packets - #/No: 18/18
      15.179908 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-06T10:49:06.573.132.000 to 2017-02-06T10:49:26.326.642.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.572735 in 1 packets - #/No: 18/18
      15.179908 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4268514 to 31.4267818
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.9999826 in 1 packets - #/No: 18/18
      14.9999478 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170206_104853_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.9999827 to 12.0002717
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.5727190  -- OK for SL --
   MIU time from 2017-02-06T10:49:06.573.132.000 to 2017-02-06T10:49:06.573.421.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.572735

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5729907 to 16.5732797
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1798550  -- OK for SL --
   MIU time from 2017-02-06T10:49:11.146.156.000 to 2017-02-06T10:49:11.146.445.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.179908

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7531347 to 31.7534237
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-06T10:49:26.326.353.000 to 2017-02-06T10:49:26.326.642.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-06T10:49:06.572.991.750
   Max SL corrected packet time: 2017-02-06T10:49:26.326.501.750
   Min SL corrected XO rel time: 10.5731332
   Max SL corrected XO rel time: 30.3266430
   TM values from -28 to 20 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-06T10:49:06.572.991.750 (XO: 10.5731332, t=35.6°C)  Rank: 00008
   TM values: Max =    19  Min =   -28  Ave =    -4.70  Std =     5.89
   Min TM value:-28 at an offset of 3.488 usec -> 2017-02-06T10:49:06.572.995.237 (XO: 10.5731367)
   Max TM value:19 at an offset of 144.113 usec -> 2017-02-06T10:49:06.573.135.862 (XO: 10.5732774)
   Max dTM/dt [1/us]::2.42e+02 at an offset of 65.225 usec -> 2017-02-06T10:49:06.573.056.975 (XO: 10.5731985)
   Min dTM/dt [1/us]::-2.26e+02 at an offset of 40.113 usec -> 2017-02-06T10:49:06.573.031.862 (XO: 10.5731734)

 - Block number: 2 starting at 2017-02-06T10:49:11.146.015.750 (XO: 15.1461572, t=35.6°C)  Rank: 00006
   TM values: Max =    20  Min =   -25  Ave =    -4.59  Std =     5.36
   Min TM value:-25 at an offset of 116.638 usec -> 2017-02-06T10:49:11.146.132.387 (XO: 15.1462738)
   Max TM value:20 at an offset of 95.888 usec -> 2017-02-06T10:49:11.146.111.637 (XO: 15.1462531)
   Max dTM/dt [1/us]::1.69e+02 at an offset of 161.251 usec -> 2017-02-06T10:49:11.146.177.000 (XO: 15.1463184)
   Min dTM/dt [1/us]::-1.62e+02 at an offset of 206.526 usec -> 2017-02-06T10:49:11.146.222.275 (XO: 15.1463637)

 - Block number: 3 starting at 2017-02-06T10:49:26.326.212.750 (XO: 30.3263540, t=35.6°C)  Rank: 00004
   TM values: Max =    16  Min =   -25  Ave =    -4.60  Std =     5.40
   Min TM value:-25 at an offset of 131.238 usec -> 2017-02-06T10:49:26.326.343.987 (XO: 30.3264852)
   Max TM value:16 at an offset of 127.400 usec -> 2017-02-06T10:49:26.326.340.150 (XO: 30.3264814)
   Max dTM/dt [1/us]::1.40e+02 at an offset of 152.976 usec -> 2017-02-06T10:49:26.326.365.725 (XO: 30.3265070)
   Min dTM/dt [1/us]::-1.55e+02 at an offset of 63.925 usec -> 2017-02-06T10:49:26.326.276.675 (XO: 30.3264179)
