-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170203_075533_1229.dat, 36720  packets found.
APID:
      1229 (EW) in 36720 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36720 packets (all packets)
Packet no from 1 to 36720
   Differences of packet numbers in neigboring packets: 
      1 in 36719 packets (all packets)
Total number of packets: 
      36720 in 36720 packets (all packets)
Hz and MHz counters from 16.084341 to 177.076374
-each separately from 16s to 177s and from 76374us to 563482us.
   Differences in neigboring packets: 
      0.000000 in 36715 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -129.512892 in 1 packets - #/No: 6324/6324
      129.512892 in 1 packets - #/No: 18564/18564
      -160.992033 in 1 packets - #/No: 18768/18768
      160.992033 in 1 packets - #/No: 31008/31008
MIU packet time from 2017-02-03T07:55:52.084.341.000 to 2017-02-03T07:58:33.076.374.000
   Differences in neigboring packets: 
      0.000000 in 36715 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -129.512892 in 1 packets - #/No: 6324/6324
      129.512892 in 1 packets - #/No: 18564/18564
      -160.992033 in 1 packets - #/No: 18768/18768
      160.992033 in 1 packets - #/No: 31008/31008
MIU orbit number: 
      60000 in 36720 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36720 packets (all packets)
MIU orbit information / number of events: 
      0 in 36720 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36720 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36720 packets (all packets)
FPGA configuration version: 
      211 in 36720 packets (all packets)
10 MHz counter at the last 1Hz pulse from 18.1071274 to 179.1065912
   Differences in neigboring packets: 
      0.0000000 in 36715 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -129.9995678 in 1 packets - #/No: 6324/6324
      129.9995678 in 1 packets - #/No: 18564/18564
      -160.9994638 in 1 packets - #/No: 18768/18768
      160.9994638 in 1 packets - #/No: 31008/31008
Temperature FPGA: 
      05 -> 48.5°C in 24480 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 12240 packets - #/No: 18769/18769, 18770/18770, 18771/18771, 18772/18772, 18773/18773, 18774/18774, 18775/18775, 18776/18776, 18777/18777, 18778/18778, 18779/18779, 18780/18780, 18781/18781, 18782/18782, 18783/18783, 18784/18784, 18785/18785, 18786/18786, ...
Temperature XO: 
      C0 -> 35.6°C in 36720 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 36720 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 36720 packets (all packets)
Number of external events: 
      3 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      2 in 12240 packets - #/No: 6325/6325, 6326/6326, 6327/6327, 6328/6328, 6329/6329, 6330/6330, 6331/6331, 6332/6332, 6333/6333, 6334/6334, 6335/6335, 6336/6336, 6337/6337, 6338/6338, 6339/6339, 6340/6340, 6341/6341, 6342/6342, ...
      1 in 12240 packets - #/No: 18769/18769, 18770/18770, 18771/18771, 18772/18772, 18773/18773, 18774/18774, 18775/18775, 18776/18776, 18777/18777, 18778/18778, 18779/18779, 18780/18780, 18781/18781, 18782/18782, 18783/18783, 18784/18784, 18785/18785, 18786/18786, ...
Number of generated alerts: 
      4 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      3 in 12240 packets - #/No: 6325/6325, 6326/6326, 6327/6327, 6328/6328, 6329/6329, 6330/6330, 6331/6331, 6332/6332, 6333/6333, 6334/6334, 6335/6335, 6336/6336, 6337/6337, 6338/6338, 6339/6339, 6340/6340, 6341/6341, 6342/6342, ...
      2 in 12240 packets - #/No: 18769/18769, 18770/18770, 18771/18771, 18772/18772, 18773/18773, 18774/18774, 18775/18775, 18776/18776, 18777/18777, 18778/18778, 18779/18779, 18780/18780, 18781/18781, 18782/18782, 18783/18783, 18784/18784, 18785/18785, 18786/18786, ...
Survey memory error: 
      0 in 36720 packets (all packets)
Event memory error: 
      0 in 36720 packets (all packets)
Configuration error: 
      0 in 36720 packets (all packets)
Zero: 
      0 in 36720 packets (all packets)
Radio gain: 
      0 in 36720 packets (all packets)
Radio bandwidth: 
      0 in 36720 packets (all packets)
Digital gain: 
      0 in 36720 packets (all packets)
Analog gain: 
      0 in 36720 packets (all packets)

EVENT DATA READER
-- File 20170203_075533_1229.dat, 180 EW pages found. Extra packets: 0.
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      3 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      2 in 12240 packets - #/No: 6325/6325, 6326/6326, 6327/6327, 6328/6328, 6329/6329, 6330/6330, 6331/6331, 6332/6332, 6333/6333, 6334/6334, 6335/6335, 6336/6336, 6337/6337, 6338/6338, 6339/6339, 6340/6340, 6341/6341, 6342/6342, ...
      1 in 12240 packets - #/No: 18769/18769, 18770/18770, 18771/18771, 18772/18772, 18773/18773, 18774/18774, 18775/18775, 18776/18776, 18777/18777, 18778/18778, 18779/18779, 18780/18780, 18781/18781, 18782/18782, 18783/18783, 18784/18784, 18785/18785, 18786/18786, ...
   10 MHz counter from 18.1359814 to 179.3355404
      Difference between the neighboring packets: 
      0.0000170 in 36714 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -0.2080630 in 1 packets - #/No: 5712/5712
      -129.5228470 in 1 packets - #/No: 6324/6324
      129.3148010 in 1 packets - #/No: 18564/18564
      -161.0053510 in 1 packets - #/No: 18768/18768
      160.7973050 in 1 packets - #/No: 31008/31008
   Memory page number from 0 to 237
      Difference between the neighboring packets: 
      0 in 36540 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 178 packets - #/No: 204/204, 408/408, 612/612, 816/816, 1020/1020, 1224/1224, 1428/1428, 1632/1632, 1836/1836, 2040/2040, 2244/2244, 2448/2448, 2652/2652, 2856/2856, 3060/3060, 3264/3264, 3468/3468, 3672/3672, ...
      59 in 1 packets - #/No: 5712/5712
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 36540 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 174 packets - #/No: 204/204, 408/408, 612/612, 816/816, 1020/1020, 1224/1224, 1428/1428, 1632/1632, 1836/1836, 2040/2040, 2244/2244, 2448/2448, 2652/2652, 2856/2856, 3060/3060, 3264/3264, 3468/3468, 3672/3672, ...
      -59 in 1 packets - #/No: 5712/5712
      -2 in 1 packets - #/No: 6324/6324
      -56 in 1 packets - #/No: 18564/18564
      -3 in 1 packets - #/No: 18768/18768
      -55 in 1 packets - #/No: 31008/31008
   MIU event flag 
      1 in 36720 packets (all packets)
   Time triggered event flag 
      0 in 36720 packets (all packets)
   Min EW event (1st sample) abs. time: 2017-02-03T07:55:52.028.852.562
   Min EW header abs. time            : 2017-02-03T07:55:52.084.341.000
   Max EW event (1st sample) abs. time: 2017-02-03T07:58:33.020.885.562
   Max EW header abs. time            : 2017-02-03T07:58:33.076.374.000
   TM values from -747 to 537 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      150 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      151 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      152 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      153 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      154 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      155 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      156 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      157 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      158 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      159 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      160 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      161 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      162 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      163 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      164 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      165 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      166 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      167 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      168 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      169 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      170 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      171 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      172 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      173 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      174 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      175 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      176 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      177 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      178 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      179 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      180 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      181 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      182 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      183 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      184 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      185 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      186 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      187 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      188 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      189 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      190 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      191 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      192 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      193 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      194 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      195 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      196 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      197 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      198 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      199 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      200 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      201 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      202 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      203 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      204 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      205 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      206 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      207 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      208 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      209 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 60  ordered packets: 12240
   TM values from -747 to 537 median:-4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=16 C1MHz=0.084341
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=18.1359814
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2017-02-03T07:55:52.028.852.562
      (header abs. time: 2017-02-03T07:55:52.084.341.000)
      Event 1st sample XO time (corr. sec. from XP start):              16.028853848
      Event header XO time (corr. sec. from XP start)    :              16.028854098
      Sampling frequency correction factor for this event:     0.999996600

 Event no. 2 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      106 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      107 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      108 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      109 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      110 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      111 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      112 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      113 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      114 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      115 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      116 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      117 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      118 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      119 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      120 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      121 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      122 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      123 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      124 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      125 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      126 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      127 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      128 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      129 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      130 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      131 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      132 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      133 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      134 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      135 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      136 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      137 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      138 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      139 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      140 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      141 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      142 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      143 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      144 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      145 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      146 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      147 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      148 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 60  ordered packets: 12240
   TM values from -498 to 282 median:-5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=47 C1MHz=0.563482
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=49.6150174
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2017-02-03T07:56:23.507.993.562
      (header abs. time: 2017-02-03T07:56:23.563.482.000)
      Event 1st sample XO time (corr. sec. from XP start):              47.507997577
      Event header XO time (corr. sec. from XP start)    :              47.507997827
      Sampling frequency correction factor for this event:     0.999996600

 Event no. 3 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      149 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      210 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      211 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      212 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      213 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      214 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      215 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      216 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      217 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      218 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      219 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      220 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      221 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      222 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      223 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      224 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      225 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      226 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      227 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      228 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      229 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      230 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      231 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      232 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      233 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      234 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      235 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      236 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      237 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 60  ordered packets: 12240
   TM values from -317 to 270 median:-5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=177 C1MHz=0.076374
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=179.1274774
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2017-02-03T07:58:33.020.885.562
      (header abs. time: 2017-02-03T07:58:33.076.374.000)
      Event 1st sample XO time (corr. sec. from XP start):             177.020897921
      Event header XO time (corr. sec. from XP start)    :             177.020898171
      Sampling frequency correction factor for this event:     0.999996600
--
Extremes in separate events:

 - Event number: 1 starting at 2017-02-03T07:55:52.028.852.562 (XO: 16.0288538)
   Min TM value:-747 at an offset of 54705.586 usec -> 2017-02-03T07:55:52.083.558.147 (XO: 16.0835594)
   Max TM value:537 at an offset of 54717.886 usec -> 2017-02-03T07:55:52.083.570.448 (XO: 16.0835717)
   Max dTM/dt [1/us]:3.40e+03 at an offset of 54705.661 usec -> 2017-02-03T07:55:52.083.558.222 (XO: 16.0835595)
   Min dTM/dt [1/us]:-4.28e+03 at an offset of 54714.799 usec -> 2017-02-03T07:55:52.083.567.360 (XO: 16.0835686)

 - Event number: 2 starting at 2017-02-03T07:56:23.507.993.562 (XO: 47.5079976)
   Min TM value:-498 at an offset of 51961.689 usec -> 2017-02-03T07:56:23.559.955.251 (XO: 47.5599593)
   Max TM value:282 at an offset of 51962.002 usec -> 2017-02-03T07:56:23.559.955.563 (XO: 47.5599596)
   Max dTM/dt [1/us]:4.68e+03 at an offset of 52169.890 usec -> 2017-02-03T07:56:23.560.163.451 (XO: 47.5601675)
   Min dTM/dt [1/us]:-3.96e+03 at an offset of 51961.639 usec -> 2017-02-03T07:56:23.559.955.201 (XO: 47.5599592)

 - Event number: 3 starting at 2017-02-03T07:58:33.020.885.562 (XO: 177.0208979)
   Min TM value:-317 at an offset of 186983.873 usec -> 2017-02-03T07:58:33.207.869.435 (XO: 177.2078818)
   Max TM value:270 at an offset of 53235.944 usec -> 2017-02-03T07:58:33.074.121.505 (XO: 177.0741339)
   Max dTM/dt [1/us]:3.52e+03 at an offset of 191748.002 usec -> 2017-02-03T07:58:33.212.633.563 (XO: 177.2126459)
   Min dTM/dt [1/us]:-2.67e+03 at an offset of 190978.199 usec -> 2017-02-03T07:58:33.211.863.761 (XO: 177.2118761)
