-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170130_083838_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.811954 to 30.550229
-each separately from 1s to 30s and from 549940us to 873499us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.060967 in 1 packets - #/No: 18/18
      2.676441 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-30T08:38:42.811.954.000 to 2017-01-30T08:39:11.550.229.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.060967 in 1 packets - #/No: 18/18
      2.676441 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9302208 to 31.9301218
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      25.9999113 in 1 packets - #/No: 18/18
      2.9999897 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170130_083838_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7421735 to 3.7424625
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.0608790  -- OK for SL --
   MIU time from 2017-01-30T08:38:42.811.954.000 to 2017-01-30T08:38:42.812.243.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.060967

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.8033415 to 29.8036305
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.6764310  -- OK for SL --
   MIU time from 2017-01-30T08:39:08.873.210.000 to 2017-01-30T08:39:08.873.499.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.676441

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.4800615 to 32.4803505
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-30T08:39:11.549.940.000 to 2017-01-30T08:39:11.550.229.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-30T08:38:42.811.813.750
   Max SL corrected packet time: 2017-01-30T08:39:11.550.088.750
   Min SL corrected XO rel time: 1.8119555
   Max SL corrected XO rel time: 30.5502306
   TM values from -30 to 23 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-30T08:38:42.811.813.750 (XO: 1.8119555, t=35.6°C)  Rank: 00005
   TM values: Max =    14  Min =   -22  Ave =    -4.50  Std =     4.70
   Min TM value:-22 at an offset of 123.850 usec -> 2017-01-30T08:38:42.811.937.600 (XO: 1.8120793)
   Max TM value:14 at an offset of 98.725 usec -> 2017-01-30T08:38:42.811.912.475 (XO: 1.8120542)
   Max dTM/dt [1/us]::1.32e+02 at an offset of 162.201 usec -> 2017-01-30T08:38:42.811.975.950 (XO: 1.8121177)
   Min dTM/dt [1/us]::-1.22e+02 at an offset of 98.963 usec -> 2017-01-30T08:38:42.811.912.712 (XO: 1.8120544)

 - Block number: 2 starting at 2017-01-30T08:39:08.873.069.750 (XO: 27.8732124, t=35.6°C)  Rank: 00007
   TM values: Max =    23  Min =   -30  Ave =    -4.50  Std =     7.66
   Min TM value:-30 at an offset of 116.525 usec -> 2017-01-30T08:39:08.873.186.275 (XO: 27.8733289)
   Max TM value:23 at an offset of 127.863 usec -> 2017-01-30T08:39:08.873.197.612 (XO: 27.8733402)
   Max dTM/dt [1/us]::1.90e+02 at an offset of 120.013 usec -> 2017-01-30T08:39:08.873.189.762 (XO: 27.8733324)
   Min dTM/dt [1/us]::-1.96e+02 at an offset of 155.876 usec -> 2017-01-30T08:39:08.873.225.625 (XO: 27.8733682)

 - Block number: 3 starting at 2017-01-30T08:39:11.549.799.750 (XO: 30.5499416, t=35.6°C)  Rank: 00007
   TM values: Max =    16  Min =   -24  Ave =    -4.39  Std =     6.41
   Min TM value:-24 at an offset of 122.638 usec -> 2017-01-30T08:39:11.549.922.387 (XO: 30.5500642)
   Max TM value:16 at an offset of 132.425 usec -> 2017-01-30T08:39:11.549.932.175 (XO: 30.5500740)
   Max dTM/dt [1/us]::1.82e+02 at an offset of 164.763 usec -> 2017-01-30T08:39:11.549.964.513 (XO: 30.5501063)
   Min dTM/dt [1/us]::-1.87e+02 at an offset of 156.988 usec -> 2017-01-30T08:39:11.549.956.738 (XO: 30.5500986)
