-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170126_175044_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.978964 to 30.498775
-each separately from 0s to 30s and from 346036us to 979253us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.366783 in 1 packets - #/No: 18/18
      15.152161 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-26T17:50:47.978.964.000 to 2017-01-26T17:51:17.498.775.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.366783 in 1 packets - #/No: 18/18
      15.152161 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4298120 to 31.4297123
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999502 in 1 packets - #/No: 18/18
      14.9999501 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170126_175044_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4087742 to 2.4090632
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3667350  -- OK for SL --
   MIU time from 2017-01-26T17:50:47.978.964.000 to 2017-01-26T17:50:47.979.253.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.366783

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7757982 to 16.7760872
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1521110  -- OK for SL --
   MIU time from 2017-01-26T17:51:02.346.036.000 to 2017-01-26T17:51:02.346.325.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.152161

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9281982 to 31.9284872
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-26T17:51:17.498.486.000 to 2017-01-26T17:51:17.498.775.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-26T17:50:47.978.823.750
   Max SL corrected packet time: 2017-01-26T17:51:17.498.634.750
   Min SL corrected XO rel time: 0.9789654
   Max SL corrected XO rel time: 30.4987765
   TM values from -45 to 30 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-26T17:50:47.978.823.750 (XO: 0.9789654, t=35.6°C)  Rank: 00007
   TM values: Max =    28  Min =   -38  Ave =    -4.48  Std =     8.52
   Min TM value:-38 at an offset of 110.313 usec -> 2017-01-26T17:50:47.978.934.062 (XO: 0.9790757)
   Max TM value:28 at an offset of 111.200 usec -> 2017-01-26T17:50:47.978.934.950 (XO: 0.9790766)
   Max dTM/dt [1/us]::3.89e+02 at an offset of 116.563 usec -> 2017-01-26T17:50:47.978.940.312 (XO: 0.9790820)
   Min dTM/dt [1/us]::-4.15e+02 at an offset of 133.363 usec -> 2017-01-26T17:50:47.978.957.112 (XO: 0.9790988)

 - Block number: 2 starting at 2017-01-26T17:51:02.345.895.750 (XO: 15.3460371, t=35.6°C)  Rank: 00007
   TM values: Max =    27  Min =   -39  Ave =    -4.44  Std =     8.64
   Min TM value:-39 at an offset of 140.513 usec -> 2017-01-26T17:51:02.346.036.262 (XO: 15.3461777)
   Max TM value:27 at an offset of 127.625 usec -> 2017-01-26T17:51:02.346.023.375 (XO: 15.3461648)
   Max dTM/dt [1/us]::4.04e+02 at an offset of 50.825 usec -> 2017-01-26T17:51:02.345.946.575 (XO: 15.3460880)
   Min dTM/dt [1/us]::-3.75e+02 at an offset of 247.001 usec -> 2017-01-26T17:51:02.346.142.750 (XO: 15.3462841)

 - Block number: 3 starting at 2017-01-26T17:51:17.498.345.750 (XO: 30.4984875, t=35.6°C)  Rank: 00007
   TM values: Max =    30  Min =   -45  Ave =    -4.29  Std =    10.16
   Min TM value:-45 at an offset of 72.325 usec -> 2017-01-26T17:51:17.498.418.075 (XO: 30.4985599)
   Max TM value:30 at an offset of 183.901 usec -> 2017-01-26T17:51:17.498.529.650 (XO: 30.4986714)
   Max dTM/dt [1/us]::4.10e+02 at an offset of 146.813 usec -> 2017-01-26T17:51:17.498.492.562 (XO: 30.4986344)
   Min dTM/dt [1/us]::-4.25e+02 at an offset of 211.938 usec -> 2017-01-26T17:51:17.498.557.688 (XO: 30.4986995)
