-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170124_030856_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.971941 to 36.142049
-each separately from 8s to 36s and from 141760us to 972230us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.480941 in 1 packets - #/No: 18/18
      13.688300 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-24T03:09:07.971.941.000 to 2017-01-24T03:09:35.142.049.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.480941 in 1 packets - #/No: 18/18
      13.688300 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4239599 to 37.4238664
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999533 in 1 packets - #/No: 18/18
      13.9999532 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170124_030856_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.3958996 to 10.3961886
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4808950  -- OK for SL --
   MIU time from 2017-01-24T03:09:07.971.941.000 to 2017-01-24T03:09:07.972.230.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.480941

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.8770836 to 23.8773726
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6882550  -- OK for SL --
   MIU time from 2017-01-24T03:09:21.453.171.000 to 2017-01-24T03:09:21.453.460.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.688300

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.5656276 to 37.5659166
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-24T03:09:35.141.760.000 to 2017-01-24T03:09:35.142.049.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-24T03:09:07.971.800.750
   Max SL corrected packet time: 2017-01-24T03:09:35.141.908.750
   Min SL corrected XO rel time: 8.9719429
   Max SL corrected XO rel time: 36.1420507
   TM values from -38 to 33 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-24T03:09:07.971.800.750 (XO: 8.9719429, t=34.0°C)  Rank: 00008
   TM values: Max =    26  Min =   -32  Ave =    -4.31  Std =     7.71
   Min TM value:-32 at an offset of 99.663 usec -> 2017-01-24T03:09:07.971.900.412 (XO: 8.9720426)
   Max TM value:26 at an offset of 175.488 usec -> 2017-01-24T03:09:07.971.976.238 (XO: 8.9721184)
   Max dTM/dt [1/us]::2.64e+02 at an offset of 42.875 usec -> 2017-01-24T03:09:07.971.843.625 (XO: 8.9719858)
   Min dTM/dt [1/us]::-2.66e+02 at an offset of 275.788 usec -> 2017-01-24T03:09:07.972.076.538 (XO: 8.9722187)

 - Block number: 2 starting at 2017-01-24T03:09:21.453.030.750 (XO: 22.4531719, t=34.0°C)  Rank: 00005
   TM values: Max =    23  Min =   -29  Ave =    -4.40  Std =     6.50
   Min TM value:-29 at an offset of 100.888 usec -> 2017-01-24T03:09:21.453.131.637 (XO: 22.4532728)
   Max TM value:23 at an offset of 94.488 usec -> 2017-01-24T03:09:21.453.125.237 (XO: 22.4532664)
   Max dTM/dt [1/us]::3.30e+02 at an offset of 194.763 usec -> 2017-01-24T03:09:21.453.225.513 (XO: 22.4533667)
   Min dTM/dt [1/us]::-3.46e+02 at an offset of 147.351 usec -> 2017-01-24T03:09:21.453.178.100 (XO: 22.4533193)

 - Block number: 3 starting at 2017-01-24T03:09:35.141.619.750 (XO: 36.1417617, t=34.0°C)  Rank: 00010
   TM values: Max =    33  Min =   -38  Ave =    -4.43  Std =     8.52
   Min TM value:-38 at an offset of 212.363 usec -> 2017-01-24T03:09:35.141.832.113 (XO: 36.1419740)
   Max TM value:33 at an offset of 116.763 usec -> 2017-01-24T03:09:35.141.736.512 (XO: 36.1418784)
   Max dTM/dt [1/us]::2.90e+02 at an offset of 212.051 usec -> 2017-01-24T03:09:35.141.831.800 (XO: 36.1419737)
   Min dTM/dt [1/us]::-3.08e+02 at an offset of 212.288 usec -> 2017-01-24T03:09:35.141.832.038 (XO: 36.1419740)
