-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170119_124059_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.702138 to 36.480880
-each separately from 4s to 36s and from 480591us to 702427us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.899010 in 1 packets - #/No: 18/18
      20.878865 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-19T12:41:06.702.138.000 to 2017-01-19T12:41:38.480.880.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.899010 in 1 packets - #/No: 18/18
      20.878865 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.4237900 to 37.4236880
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.9999648 in 1 packets - #/No: 18/18
      20.9999332 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170119_124059_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.1259268 to 6.1262158
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8989750  -- OK for SL --
   MIU time from 2017-01-19T12:41:06.702.138.000 to 2017-01-19T12:41:06.702.427.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.899010

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0251908 to 17.0254798
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.8787990  -- OK for SL --
   MIU time from 2017-01-19T12:41:17.601.437.000 to 2017-01-19T12:41:17.601.726.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.878865

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.9042788 to 37.9045678
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-19T12:41:38.480.591.000 to 2017-01-19T12:41:38.480.880.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-19T12:41:06.701.997.750
   Max SL corrected packet time: 2017-01-19T12:41:38.480.739.750
   Min SL corrected XO rel time: 4.7021391
   Max SL corrected XO rel time: 36.4808813
   TM values from -31 to 21 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-19T12:41:06.701.997.750 (XO: 4.7021391, t=34.0°C)  Rank: 00006
   TM values: Max =    21  Min =   -31  Ave =    -4.62  Std =     9.50
   Min TM value:-31 at an offset of 53.388 usec -> 2017-01-19T12:41:06.702.051.137 (XO: 4.7021925)
   Max TM value:21 at an offset of 55.625 usec -> 2017-01-19T12:41:06.702.053.375 (XO: 4.7021947)
   Max dTM/dt [1/us]::2.28e+02 at an offset of 305.989 usec -> 2017-01-19T12:41:06.702.303.738 (XO: 4.7024451)
   Min dTM/dt [1/us]::-1.95e+02 at an offset of 153.663 usec -> 2017-01-19T12:41:06.702.151.413 (XO: 4.7022928)

 - Block number: 2 starting at 2017-01-19T12:41:17.601.296.750 (XO: 15.6014379, t=34.0°C)  Rank: 00005
   TM values: Max =    17  Min =   -26  Ave =    -4.70  Std =     6.21
   Min TM value:-26 at an offset of 118.300 usec -> 2017-01-19T12:41:17.601.415.050 (XO: 15.6015562)
   Max TM value:17 at an offset of 119.175 usec -> 2017-01-19T12:41:17.601.415.925 (XO: 15.6015571)
   Max dTM/dt [1/us]::1.52e+02 at an offset of 94.500 usec -> 2017-01-19T12:41:17.601.391.250 (XO: 15.6015324)
   Min dTM/dt [1/us]::-1.65e+02 at an offset of 127.238 usec -> 2017-01-19T12:41:17.601.423.987 (XO: 15.6015652)

 - Block number: 3 starting at 2017-01-19T12:41:38.480.450.750 (XO: 36.4805923, t=34.0°C)  Rank: 00006
   TM values: Max =    14  Min =   -23  Ave =    -4.74  Std =     5.35
   Min TM value:-23 at an offset of 17.513 usec -> 2017-01-19T12:41:38.480.468.262 (XO: 36.4806098)
   Max TM value:14 at an offset of 56.113 usec -> 2017-01-19T12:41:38.480.506.862 (XO: 36.4806484)
   Max dTM/dt [1/us]::1.85e+02 at an offset of 70.938 usec -> 2017-01-19T12:41:38.480.521.687 (XO: 36.4806632)
   Min dTM/dt [1/us]::-2.36e+02 at an offset of 38.150 usec -> 2017-01-19T12:41:38.480.488.900 (XO: 36.4806304)
