-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170116_102506_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.175007 to 30.457205
-each separately from 0s to 30s and from 175007us to 674164us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.498579 in 1 packets - #/No: 18/18
      14.782752 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-16T10:25:09.175.007.000 to 2017-01-16T10:25:39.457.205.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.498579 in 1 packets - #/No: 18/18
      14.782752 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4205703 to 31.4204702
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999500 in 1 packets - #/No: 18/18
      14.9999499 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170116_102506_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5955784 to 1.5958674
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4985270  -- OK for SL --
   MIU time from 2017-01-16T10:25:09.175.007.000 to 2017-01-16T10:25:09.175.296.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.498579

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0943944 to 17.0946834
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7827030  -- OK for SL --
   MIU time from 2017-01-16T10:25:24.673.875.000 to 2017-01-16T10:25:24.674.164.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.782752

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.8773864 to 31.8776754
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-16T10:25:39.456.916.000 to 2017-01-16T10:25:39.457.205.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-16T10:25:09.174.866.750
   Max SL corrected packet time: 2017-01-16T10:25:39.457.064.750
   Min SL corrected XO rel time: 0.1750087
   Max SL corrected XO rel time: 30.4572067
   TM values from -22 to 12 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-16T10:25:09.174.866.750 (XO: 0.1750087, t=34.0°C)  Rank: 00004
   TM values: Max =    11  Min =   -22  Ave =    -4.65  Std =     5.15
   Min TM value:-22 at an offset of 121.675 usec -> 2017-01-16T10:25:09.174.988.425 (XO: 0.1751304)
   Max TM value:11 at an offset of 82.763 usec -> 2017-01-16T10:25:09.174.949.512 (XO: 0.1750915)
   Max dTM/dt [1/us]::1.09e+02 at an offset of 135.800 usec -> 2017-01-16T10:25:09.175.002.550 (XO: 0.1751445)
   Min dTM/dt [1/us]::-1.03e+02 at an offset of 165.413 usec -> 2017-01-16T10:25:09.175.032.163 (XO: 0.1751741)

 - Block number: 2 starting at 2017-01-16T10:25:24.673.734.750 (XO: 15.6738764, t=34.0°C)  Rank: 00004
   TM values: Max =    12  Min =   -21  Ave =    -4.60  Std =     4.75
   Min TM value:-21 at an offset of 133.938 usec -> 2017-01-16T10:25:24.673.868.687 (XO: 15.6740103)
   Max TM value:12 at an offset of 91.888 usec -> 2017-01-16T10:25:24.673.826.637 (XO: 15.6739683)
   Max dTM/dt [1/us]::1.02e+02 at an offset of 156.976 usec -> 2017-01-16T10:25:24.673.891.725 (XO: 15.6740334)
   Min dTM/dt [1/us]::-1.03e+02 at an offset of 142.688 usec -> 2017-01-16T10:25:24.673.877.437 (XO: 15.6740191)

 - Block number: 3 starting at 2017-01-16T10:25:39.456.775.750 (XO: 30.4569177, t=34.0°C)  Rank: 00004
   TM values: Max =    11  Min =   -21  Ave =    -4.59  Std =     5.17
   Min TM value:-21 at an offset of 159.426 usec -> 2017-01-16T10:25:39.456.935.175 (XO: 30.4570771)
   Max TM value:11 at an offset of 240.876 usec -> 2017-01-16T10:25:39.457.016.625 (XO: 30.4571586)
   Max dTM/dt [1/us]::9.80e+01 at an offset of 188.663 usec -> 2017-01-16T10:25:39.456.964.413 (XO: 30.4571064)
   Min dTM/dt [1/us]::-9.60e+01 at an offset of 189.726 usec -> 2017-01-16T10:25:39.456.965.475 (XO: 30.4571074)
