-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170112_150708_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.446159 to 33.042299
-each separately from 0s to 33s and from 42010us to 687000us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.240263 in 1 packets - #/No: 18/18
      15.355010 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-12T15:07:11.446.159.000 to 2017-01-12T15:07:44.042.299.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.240263 in 1 packets - #/No: 18/18
      15.355010 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4286242 to 34.4285160
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999442 in 1 packets - #/No: 18/18
      15.9999476 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170112_150708_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8747827 to 1.8750717
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.2402070  -- OK for SL --
   MIU time from 2017-01-12T15:07:11.446.159.000 to 2017-01-12T15:07:11.446.448.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.240263

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1152787 to 19.1155677
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3549590  -- OK for SL --
   MIU time from 2017-01-12T15:07:28.686.711.000 to 2017-01-12T15:07:28.687.000.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.355010

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.4705267 to 34.4708157
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-12T15:07:44.042.010.000 to 2017-01-12T15:07:44.042.299.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-12T15:07:11.446.018.750
   Max SL corrected packet time: 2017-01-12T15:07:44.042.158.750
   Min SL corrected XO rel time: 0.4461600
   Max SL corrected XO rel time: 33.0422998
   TM values from -31 to 24 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-12T15:07:11.446.018.750 (XO: 0.4461600, t=35.6°C)  Rank: 00005
   TM values: Max =    19  Min =   -27  Ave =    -4.59  Std =     6.12
   Min TM value:-27 at an offset of 123.188 usec -> 2017-01-12T15:07:11.446.141.937 (XO: 0.4462832)
   Max TM value:19 at an offset of 145.188 usec -> 2017-01-12T15:07:11.446.163.937 (XO: 0.4463052)
   Max dTM/dt [1/us]::1.92e+02 at an offset of 144.750 usec -> 2017-01-12T15:07:11.446.163.500 (XO: 0.4463047)
   Min dTM/dt [1/us]::-2.03e+02 at an offset of 145.225 usec -> 2017-01-12T15:07:11.446.163.975 (XO: 0.4463052)

 - Block number: 2 starting at 2017-01-12T15:07:28.686.570.750 (XO: 17.6867125, t=35.6°C)  Rank: 00005
   TM values: Max =    24  Min =   -29  Ave =    -4.55  Std =     6.70
   Min TM value:-29 at an offset of 127.738 usec -> 2017-01-12T15:07:28.686.698.487 (XO: 17.6868402)
   Max TM value:24 at an offset of 125.475 usec -> 2017-01-12T15:07:28.686.696.225 (XO: 17.6868380)
   Max dTM/dt [1/us]::2.61e+02 at an offset of 187.238 usec -> 2017-01-12T15:07:28.686.757.988 (XO: 17.6868997)
   Min dTM/dt [1/us]::-2.80e+02 at an offset of 89.338 usec -> 2017-01-12T15:07:28.686.660.087 (XO: 17.6868018)

 - Block number: 3 starting at 2017-01-12T15:07:44.041.869.750 (XO: 33.0420108, t=35.6°C)  Rank: 00005
   TM values: Max =    19  Min =   -31  Ave =    -4.65  Std =     7.05
   Min TM value:-31 at an offset of 112.575 usec -> 2017-01-12T15:07:44.041.982.325 (XO: 33.0421234)
   Max TM value:19 at an offset of 103.913 usec -> 2017-01-12T15:07:44.041.973.662 (XO: 33.0421147)
   Max dTM/dt [1/us]::2.16e+02 at an offset of 28.013 usec -> 2017-01-12T15:07:44.041.897.762 (XO: 33.0420388)
   Min dTM/dt [1/us]::-2.16e+02 at an offset of 27.950 usec -> 2017-01-12T15:07:44.041.897.700 (XO: 33.0420388)
