-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170108_055617_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.009101 to 38.953704
-each separately from 1s to 38s and from 9101us to 953704us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.312672 in 1 packets - #/No: 18/18
      22.631064 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-08T05:56:21.009.101.000 to 2017-01-08T05:56:58.953.704.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.312672 in 1 packets - #/No: 18/18
      22.631064 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4254003 to 39.4252805
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999515 in 1 packets - #/No: 18/18
      21.9999287 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170108_055617_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4345024 to 2.4347914
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3126230  -- OK for SL --
   MIU time from 2017-01-08T05:56:21.009.101.000 to 2017-01-08T05:56:21.009.390.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.312672

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.7474144 to 17.7477034
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6309910  -- OK for SL --
   MIU time from 2017-01-08T05:56:36.322.062.000 to 2017-01-08T05:56:36.322.351.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.631064

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.3786944 to 40.3789834
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-08T05:56:58.953.415.000 to 2017-01-08T05:56:58.953.704.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-08T05:56:21.008.960.750
   Max SL corrected packet time: 2017-01-08T05:56:58.953.563.750
   Min SL corrected XO rel time: 1.0091021
   Max SL corrected XO rel time: 38.9537060
   TM values from -23 to 15 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-08T05:56:21.008.960.750 (XO: 1.0091021, t=34.0°C)  Rank: 00005
   TM values: Max =    15  Min =   -22  Ave =    -4.51  Std =     5.95
   Min TM value:-22 at an offset of 124.950 usec -> 2017-01-08T05:56:21.009.085.700 (XO: 1.0092271)
   Max TM value:15 at an offset of 265.638 usec -> 2017-01-08T05:56:21.009.226.388 (XO: 1.0093678)
   Max dTM/dt [1/us]::1.37e+02 at an offset of 222.838 usec -> 2017-01-08T05:56:21.009.183.588 (XO: 1.0093250)
   Min dTM/dt [1/us]::-1.42e+02 at an offset of 267.688 usec -> 2017-01-08T05:56:21.009.228.438 (XO: 1.0093698)

 - Block number: 2 starting at 2017-01-08T05:56:36.321.921.750 (XO: 16.3220637, t=34.0°C)  Rank: 00005
   TM values: Max =    13  Min =   -23  Ave =    -4.46  Std =     5.39
   Min TM value:-23 at an offset of 64.450 usec -> 2017-01-08T05:56:36.321.986.200 (XO: 16.3221281)
   Max TM value:13 at an offset of 142.838 usec -> 2017-01-08T05:56:36.322.064.587 (XO: 16.3222065)
   Max dTM/dt [1/us]::1.34e+02 at an offset of 78.475 usec -> 2017-01-08T05:56:36.322.000.225 (XO: 16.3221421)
   Min dTM/dt [1/us]::-1.37e+02 at an offset of 202.738 usec -> 2017-01-08T05:56:36.322.124.488 (XO: 16.3222664)

 - Block number: 3 starting at 2017-01-08T05:56:58.953.274.750 (XO: 38.9534170, t=34.0°C)  Rank: 00005
   TM values: Max =    14  Min =   -21  Ave =    -4.62  Std =     5.31
   Min TM value:-21 at an offset of 70.363 usec -> 2017-01-08T05:56:58.953.345.112 (XO: 38.9534874)
   Max TM value:14 at an offset of 115.213 usec -> 2017-01-08T05:56:58.953.389.962 (XO: 38.9535323)
   Max dTM/dt [1/us]::1.28e+02 at an offset of 75.375 usec -> 2017-01-08T05:56:58.953.350.125 (XO: 38.9534924)
   Min dTM/dt [1/us]::-1.67e+02 at an offset of 164.213 usec -> 2017-01-08T05:56:58.953.438.963 (XO: 38.9535813)
