-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170104_033155_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.069594 to 38.528564
-each separately from 0s to 38s and from 69594us to 528564us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.035591 in 1 packets - #/No: 18/18
      10.422512 in 1 packets - #/No: 36/18
MIU packet time from 2017-01-04T03:31:58.069.594.000 to 2017-01-04T03:32:36.528.564.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.035591 in 1 packets - #/No: 18/18
      10.422512 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4279223 to 39.4278022
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      27.9999115 in 1 packets - #/No: 18/18
      9.9999684 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170104_033155_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4975172 to 1.4978062
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.0355030  -- OK for SL --
   MIU time from 2017-01-04T03:31:58.069.594.000 to 2017-01-04T03:31:58.069.883.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.035591

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.5333092 to 29.5335982
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.4224790  -- OK for SL --
   MIU time from 2017-01-04T03:32:26.105.474.000 to 2017-01-04T03:32:26.105.763.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.422512

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.9560772 to 39.9563662
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-01-04T03:32:36.528.275.000 to 2017-01-04T03:32:36.528.564.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-01-04T03:31:58.069.453.750
   Max SL corrected packet time: 2017-01-04T03:32:36.528.423.750
   Min SL corrected XO rel time: 0.0695951
   Max SL corrected XO rel time: 38.5285656
   TM values from -31 to 22 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-01-04T03:31:58.069.453.750 (XO: 0.0695951, t=35.6°C)  Rank: 00002
   TM values: Max =     9  Min =   -16  Ave =    -4.66  Std =     3.02
   Min TM value:-16 at an offset of 87.900 usec -> 2017-01-04T03:31:58.069.541.650 (XO: 0.0696830)
   Max TM value:9 at an offset of 80.713 usec -> 2017-01-04T03:31:58.069.534.462 (XO: 0.0696758)
   Max dTM/dt [1/us]::1.28e+02 at an offset of 62.625 usec -> 2017-01-04T03:31:58.069.516.375 (XO: 0.0696577)
   Min dTM/dt [1/us]::-1.19e+02 at an offset of 217.438 usec -> 2017-01-04T03:31:58.069.671.188 (XO: 0.0698126)

 - Block number: 2 starting at 2017-01-04T03:32:26.105.333.750 (XO: 28.1054757, t=35.6°C)  Rank: 00003
   TM values: Max =    11  Min =   -22  Ave =    -4.56  Std =     3.51
   Min TM value:-22 at an offset of 110.200 usec -> 2017-01-04T03:32:26.105.443.950 (XO: 28.1055859)
   Max TM value:11 at an offset of 108.263 usec -> 2017-01-04T03:32:26.105.442.012 (XO: 28.1055840)
   Max dTM/dt [1/us]::2.06e+02 at an offset of 108.213 usec -> 2017-01-04T03:32:26.105.441.962 (XO: 28.1055840)
   Min dTM/dt [1/us]::-2.45e+02 at an offset of 110.163 usec -> 2017-01-04T03:32:26.105.443.912 (XO: 28.1055859)

 - Block number: 3 starting at 2017-01-04T03:32:36.528.134.750 (XO: 38.5282766, t=35.6°C)  Rank: 00015
   TM values: Max =    22  Min =   -31  Ave =    -4.59  Std =     6.38
   Min TM value:-31 at an offset of 107.475 usec -> 2017-01-04T03:32:36.528.242.225 (XO: 38.5283841)
   Max TM value:22 at an offset of 76.625 usec -> 2017-01-04T03:32:36.528.211.375 (XO: 38.5283533)
   Max dTM/dt [1/us]::2.93e+02 at an offset of 305.413 usec -> 2017-01-04T03:32:36.528.440.163 (XO: 38.5285821)
   Min dTM/dt [1/us]::-2.76e+02 at an offset of 251.351 usec -> 2017-01-04T03:32:36.528.386.100 (XO: 38.5285280)
